Surgical hub having variable interconnectivity capabilities

ABSTRACT

A surgical hub may obtain a hub connectivity mode based on a hub connectivity control parameter. For example, the hub connectivity mode may be selected from multiple connectivity modes that may be preconfigured, dynamically updated, semi-dynamically updated, periodically updated, or preset. The hub connectivity modes may control inter-device connectivity within a network associated with a hospital, and/or communication with an external network associated with a different hospital. The surgical hub may determine whether to provide instructional information to at least one smart surgical instrument based on the hub connectivity mode. On a condition that the hub connectivity mode does not support provisioning instructional information to surgical devices, provisioning instructional information to surgical devices may be disabled. On a condition that the hub connectivity mode supports provisioning instructional information to surgical devices, the surgical hub may determine to obtain and provide instructional information to surgical devices.

CROSS-REFERENCE TO RELATED APPLICATIONS

This application is related to an application filed contemporaneously, with U.S. patent application Ser. No. 17/062,504, filed Oct. 2, 2020, entitled METHOD FOR OPERATING TIERED OPERATION MODES IN A SURGICAL SYSTEM, the contents of which is incorporated by reference herein.

BACKGROUND

Surgical systems often incorporate an imaging system, which can allow the clinician(s) to view the surgical site and/or one or more portions thereof on one or more displays such as a monitor, for example. The display(s) can be local and/or remote to a surgical theater. An imaging system can include a scope with a camera that views the surgical site and transmits the view to a display that is viewable by a clinician. Scopes include, but are not limited to, arthroscopes, angioscopes, bronchoscopes, choledochoscopcs, colonoscopes, cytoscopes, duodenoscopes, enteroscopes, esophagogastro-duodenoscopes (gastroscopes), endoscopes, laryngoscopes, nasopharyngo-neproscopes, sigmoidoscopes, thoracoscopes, ureteroscopes, and exoscopes. Imaging systems can be limited by the information that they are able to recognize and/or convey to the clinician(s). For example, certain concealed structures, physical contours, and/or dimensions within a three-dimensional space may be unrecognizable intraoperatively by certain imaging systems. Additionally, certain imaging systems may be incapable of communicating and/or conveying certain information to the clinician(s) intraoperatively.

SUMMARY

A surgical hub may be connected, wired or wireless, with various devices and servers in the operating room, in the medical facility and/or outside of the medical facility. The surgical hub may determine the hub connectivity mode based on a hub connectivity control parameter. The hub connectivity mode may be selected from multiple connectivity modes that may be preconfigured, dynamically updated, semi-dynamically updated, periodically updated, or preset. The hub connectivity modes may control inter-device connectivity within a network associated with a hospital, and/or communication with an external network associated with a different hospital, for example.

For example, the surgical hub may determine whether to disable obtaining instructional information based on the connectivity mode. Based on a determination that the current connectivity mode is a flow-through mode, the surgical hub may disable obtaining instructional information.

For example, the surgical hub may determine whether to provide instructional information to at least one smart surgical instrument based on the hub connectivity mode. On a condition that the hub connectivity mode does not support provisioning instructional information to surgical devices, provisioning instructional information to surgical devices may be disabled. On a condition that the hub connectivity mode supports provisioning instructional information to surgical devices, the surgical hub may determine to obtain and provide instructional information to surgical devices.

For example, the surgical hub may determine whether to retrieve aggregation analysis from the remote server based on the hub connectivity mode. Based on a determination that the current hub connectivity mode supports remote data aggregation and analysis, the surgical hub may generate an aggregation analysis request. The request may be generated based on the received surgical data and may be sent to a remote server. For example, the aggregation analysis request may indicate a request for recommendation on generator data associated with a particular step in a surgical procedure. In response, the surgical hub may receive an aggregation analysis response from the remote server. For example, the aggregation analysis response may include a recommendation and/or a report. The aggregation analysis response may include one or more of: an energy mode of the generator for a particular surgical procedure, a power output of the generator for a particular surgical procedure, and/or a duration of the power output of the generator for a particular surgical procedure. The aggregation analysis response may include instructional information as described herein. The surgical hub may generate and send instructional information to one or more surgical device(s) based on the received aggregation analysis response. Based on a determination that the current hub connectivity mode supports remote data aggregation analysis, the surgical huh may disable data aggregation analysis requests.

The hub connectivity control parameter(s) may include, but not limited to, systems capabilities such as hardware capability, firmware capability and/or software capability. The hub connectivity control parameter(s) may include a consumer-controlled parameter, such as a subscription level. For example, a medical facility may purchase a subscription to hub connectivity capabilities. Some subscription level(s) may provide the hub access to surgical data gathered from external systems, while others may limit the hub connectivity to internal devices.

In an example hub connectivity mode, the surgical hub may receive information from surgical instrument(s) and may send the received information to a remote server (such as a remote processing server and/or a remote database in the cloud).

In an example connectivity mode, the surgical hub may receive information from surgical instrument(s) and may send the received information to a remote server (such as a remote processing server and/or a remote database in the cloud). The surgical hub may receive information from surgical instrument(s), obtain instructional information based on the information received from the surgical instrument(s), and may send the instructional information to one or more surgical instrument(s).

In an example connectivity mode, the surgical hub may receive information from surgical instrument(s) and may send the received information to a remote server (such as a remote processing server and/or a remote database in the cloud). The surgical hub may receive information from surgical instrument(s), obtain instructional information based on the information received from the surgical instrument(s), and may send the instructional information to one or more surgical instrument(s). The surgical hub may record various surgical information and send surgical information to a remote server for archiving and/or analysis. The archived surgical information may be aggregated with information received from other surgical hub(s), and/or surgical information associated with other medical facilities. The aggregated information may be accessed to generate instructional information to one or more surgical instrument(s). In an example, the surgical communication hub may aggregate information, such as information received from smart surgical devices, information associated with multiple surgeries, surgical information and corresponding outcome associated with multiple patients. The aggregated information may be stored in a remote database. In an example, the surgical information may be aggregated at a remote server.

BRIEF DESCRIPTION OF THE DRAWINGS

FIG. 1 is a block diagram of a computer-implemented interactive surgical system, in accordance with at least one aspect of the present disclosure.

FIG. 2 is a surgical system being used to perform a surgical procedure in an operating room, in accordance with at least one aspect of the present disclosure.

FIG. 3 is a surgical hub paired with a visualization system, a robotic system, and an intelligent instrument, in accordance with at least one aspect of the present disclosure.

FIG. 4 illustrates a surgical data network comprising a modular communication hub configured to connect modular devises lowed in one or more operating theaters of healthcare facility, or any room in a healthcare facility specially equipped for surgical operations, to the cloud, in accordance with at least one aspect of the present disclosure.

FIG. 5 illustrates a computer-implemented interactive surgical system, in accordance with at least one aspect of the present disclosure.

FIG. 6 illustrates a surgical hub comprising a plurality of modules coupled to the modular control tower, in accordance with at least one aspect of the present disclosure.

FIG. 7 illustrates a logic diagram of a control system of a surgical instrument or tool, in accordance with at least one aspect of the present disclosure.

FIG. 8 illustrates a surgical instrument or tool comprising a plurality of motors which can be activated to perform various functions, in accordance with at least one aspect of the present disclosure.

FIG. 9 illustrates a diagram of a situationally aware surgical system, in accordance with at least one aspect of the present disclosure.

FIG. 10 illustrates a timeline of an illustrative surgical procedure and the inferences that the surgical hub can make from the data detected at each step in the surgical procedure, in accordance with at least one aspect of the present disclosure.

FIG. 11 is a block diagram of the computer-implemented interactive surgical system, in accordance with at least one aspect of the present disclosure.

FIG. 12 is a block diagram which illustrates the functional architecture of the computer-implemented interactive surgical system, in accordance with at least one aspect of the present disclosure.

FIG. 13 illustrates a block diagram of a computer-implemented interactive surgical system that is configured to adaptively generate control program updates for modular devices, in accordance with at least one aspect of the present disclosure.

FIG. 14 illustrates a surgical system that includes a handle having a controller and a motor, an adapter releasably coupled to the handle, and a loading unit releasably coupled to the adapter, in accordance with at least one aspect of the present disclosure.

FIG. 15A illustrates an example flow for determining a mode of operation and operating in the determined mode, in accordance with at least one aspect of the present disclosure.

FIG. 15B illustrates an example flow for changing a mode of operation, in accordance with at least one aspect of the present disclosure.

FIGS. 16A-C illustrate example huh connectivity modes.

FIG. 17 shows an example flow for operating under tiered hub communication modes.

FIGS. 18A-C show example flows for operating under tiered hub communication modes.

FIG. 19 illustrates an example interactive surgical system, in accordance with at least one aspect of the present disclosure.

FIG. 20 illustrates an example surgical supply packaged with an RFID NFC chip.

DETAILED DESCRIPTION

Applicant of the present application owns the following U.S. Patent Applications, each of which is herein incorporated by reference in its entirety:

U.S. patent application Ser. No. 15/940,654, entitled SURGICAL HUB SITUATIONAL AWARENESS, filed Mar. 29, 2018;

-   -   U.S. patent application Ser. No. 15/940,668 titled AGGREGATION         AND REPORTING OF SURGICAL HUB DMA; filed on Mar. 29, 2018;     -   U.S. patent application Ser. No. 16/209,478, entitled METHOD FOR         SITUATIONAL AWARENESS FOR SURGICAL NETWORK OR SURGICAL NETWORK         CONNECTED DEVICE CAPABLE OF ADJUSTING FUNCTION BASED ON A SENSED         SITUATION OR USAGE, filed Dec. 4, 2018;     -   U.S. patent application Ser. No. 16/182,246, entitled         ADJUSTMENTS BASED ON AIRBORNE PARTICLE PROPERTIES, filed Nov. 6,         2018; and     -   U.S. patent application Ser. No. 16/209,385, titled METHOD OF         HUB COMMUNICATION, PROCESSING, STORAGE AND DISPLAY, filed Dec.         4, 2018.

Referring to FIG. 1 , a computer-implemented interactive surgical system 100 may include one or more surgical systems 102 and a cloud-based system (e.g., the cloud 104 that may include a remote server 113 coupled to a storage device 105). Each surgical system 102 may include at least one surgical hub 106 in communication with the cloud 104 that may include a remote server 113. In one example, as illustrated in FIG. 1 , the surgical system 102 includes a visualization system 108, a robotic system 110, and a handheld intelligent surgical instrument 112, which are configured to communicate with one another and/or the hub 106. In some aspects, a surgical system 102 may include an M number of hubs 106, an N number of visualization systems 108, an O number of robotic systems 110, and a P number of handheld intelligent surgical instruments 112, where M, N, O, and P may be integers greater than or equal to one.

In various aspects, the visualization system 108 may include one or more imaging sensors, one or more image-processing units, one or more storage arrays, and one or more displays that are strategically arranged with respect to the sterile field, as illustrated in FIG. 2 . In one aspect, the visualization system 108 may include an interface for HL7, PACS, and EMR. Various components of the visualization system 108 are described under the heading “Advanced Imaging Acquisition Module” in US. Patent Application Publication No. US 2019-0200844 A1 (U.S. patent application Ser. No. 16/209,385), titled METHOD OF HUB COMMUNICATION, PROCESSING, STORAGE AND DISPLAY, filed Dec. 4, 2018, the disclosure of which is herein incorporated by reference in its entirety.

As illustrated in FIG. 2 , a primary display 119 is positioned in the sterile field to be visible to an operator at the operating table 114. In addition, a visualization tower 111 is positioned outside the sterile field. The visualization tower 111 may include a first non-sterile display 107 and a second non-sterile display 109, which face away from each other. The visualization system 108, guided by the hub 106, is configured to utilize the displays 107, 109, and 119 to coordinate information flow to operators inside and outside the sterile field. For example, the hub 106 may cause the visualization system 108 to display a snapshot of a surgical site, as recorded by an imaging device 124, on a non-sterile display 107 or 109, while maintaining a live feed of the surgical site on the primary display 119. The snapshot on the non-sterile display 107 or 109 can permit a non-sterile operator to perform a diagnostic step relevant to the surgical procedure, for example.

In one aspect, the hub 106 may also be configured to route a diagnostic input or feedback entered by a non-sterile operator at the visualization tower 111 to the primary display 119 within the sterile field, where it can be viewed by a sterile operator at the operating table. In one example, the input can be in the form of a modification to the snapshot displayed on the non-sterile display 107 or 109, which can be routed to the primary display 119 by the hub 106.

Referring to FIG. 2 , a surgical instrument 112 is being used in the surgical procedure as part of the surgical system 102. The hub 106 may also be configured to coordinate information flow to a display of the surgical instrument 112. For example, in U.S. Patent Application Publication No. US 2019-0200844 A1 (U.S. patent application Ser. No. 16/209,385), titled METHOD OF HUB COMMUNICATION, PROCESSING, STORAGE AND DISPLAY, filed Dec. 4, 2018, the disclosure of which is herein incorporated by reference in its entirety. A diagnostic input or feedback entered by a non-sterile operator at the visualization tower 111 can be routed by the hub 106 to the surgical instrument display 115 within the sterile field, where it can be viewed by the operator of the surgical instrument 112. Example surgical instruments that are suitable for use with the surgical system 102 are described under the heading “Surgical Instrument Hardware” and in U.S. Patent Application Publication No. US 2019-0200844 A1 (U.S. patent application Ser. No. 16/209,385), titled METHOD OF HUB COMMUNICATION, PROCESSING, STORAGE AND DISPLAY, filed Dec. 4, 2018, the disclosure of which is herein incorporated by reference in its entirety, for example.

FIG. 2 depicts an example of a surgical system 102 being used to perform a surgical procedure on a patient who is lying down on an operating table 114 in a surgical operating room 116. A robotic system 110 may be used in the surgical procedure as a part of the surgical system 102. The robotic system 110 may include a surgeon's console 118, a patient side cart 120 (surgical robot), and a surgical robotic hub 122. The patient side cart 120 can manipulate at least one removably coupled surgical tool 117 through a minimally invasive incision in the body of the patient while the surgeon views the surgical site through the surgeon's console 118. An image of the surgical site can be obtained by a medical imaging device 124, which can be manipulated by the patient side cart 120 to orient the imaging device 124. The robotic hub 122 can be used to process the images of the surgical site for subsequent display to the surgeon through the surgeon's console 118.

Other types of robotic systems can be readily adapted for use with the surgical system 102. Various examples of robotic systems and surgical tools that are suitable for use with the present disclosure are described in U.S. Patent Application Publication No. US 2019-0201137 A1 (U.S. patent application Ser. No. 16/209,407), titled METHOD OF ROBOTIC HUB COMMUNICATION, DETECTION, AND CONTROL, filed Dec. 4, 2018, the disclosure of which is herein incorporated by reference in its entirety.

Various examples of cloud-based analytics that are performed by the cloud 104, and are suitable for use with the present disclosure, are described in U.S. Patent Application Publication No. US 2019-0206569 A1 (U.S. patent application Ser. No. 16/209,403), titled METHOD OF CLOUD BASED DATA ANALYTICS FOR USE WITH THE HUB, filed Dec. 4, 2018, the disclosure of which is herein incorporated by reference in its entirety.

In various aspects, the imaging device 124 may include at least one image sensor and one or more optical components. Suitable image sensors may include, but are not limited to, Charge-Coupled Device (CCD) sensors and Complementary Metal-Oxide Semiconductor (CMOS) sensors.

The optical components of the imaging device 124 may include one or more illumination sources and/or one or more lenses. The one or more illumination sources may be directed to illuminate portions of the surgical field. The one or more image sensors may receive light reflected or refracted from the surgical field, including light reflected or refracted from tissue and/or surgical instruments.

The one or more illumination sources may be configured to radiate electromagnetic energy in the visible spectrum as well as the invisible spectrum. The visible spectrum, sometimes referred to as the optical spectrum or luminous spectrum, is that portion of the electromagnetic spectrum that is visible to (i.e., can be detected by) the human eye and may be referred to as visible light or simply light. A typical human eye will respond to wavelengths in air that are from about 380 nm to about 750 nm.

The invisible spectrum (e.g., the non-luminous spectrum) is that portion of the electromagnetic spectrum that lies below and above the visible spectrum (i.e., wavelengths below about 380 nm and above about 750 nm). The invisible spectrum is not detectable by the human eye. Wavelengths greater than about 750 nm are longer than the red visible spectrum, and they become invisible infrared (IR), microwave, and radio electromagnetic radiation. Wavelengths less than about 380 nm are shorter than the violet spectrum, and they become invisible ultraviolet, x-ray, and gamma ray electromagnetic radiation.

In various aspects, the imaging device 124 is configured for use in a minimally invasive procedure. Examples of imaging devices suitable for use with the present disclosure include, but not limited to, an arthroscope, angioscope, bronchoscope, choledochoscope, colonoscope, cytoscope, duodenoscope, enteroscope, esophagogastro-duodenoscope (gastroscope), endoscope, laryngoscope, nasopharyngo-neproscope, sigmoidoscope, thoracoscope, and ureteroscope.

The imaging device may employ multi-spectrum monitoring to discriminate topography and underlying structures. A multi-spectral image is one that captures image data within specific wavelength ranges across the electromagnetic spectrum. The wavelengths may be separated by filters or by the use of instruments that are sensitive to particular wavelengths, including light from frequencies beyond the visible light range, e.g., IR and ultraviolet.

Spectral imaging can allow extraction of additional information the human eye fails to capture with its receptors for red, green, and blue. The use of multi-spectral imaging is described in greater detail under the heading “Advanced Imaging Acquisition Module” in .S. Patent Application Publication No. US 2019-0200844 A1 (U.S. patent application Ser. No. 16/209,385), titled METHOD OF HUB COMMUNICATION, PROCESSING, STORAGE AND DISPLAY, filed Dec. 4, 2018, the disclosure of which is herein incorporated by reference in its entirety. Multi-spectrum monitoring can be a useful tool in relocating a surgical field after a surgical task is completed to perform one or more of the previously described tests on the treated tissue. It is axiomatic that strict sterilization of the operating room and surgical equipment is required during any surgery. The strict hygiene and sterilization conditions required in a “surgical theater,” i.e., an operating or treatment room, necessitate the highest possible sterility of all medical devices and equipment. Part of that sterilization process is the need to sterilize anything that comes in contact with the patient or penetrates the sterile field, including the imaging device 124 and its attachments and components. It will be appreciated that the sterile field may be considered a specified area, such as within a tray or on a sterile towel, that is considered free of microorganisms, or the sterile field may be considered an area, immediately around a patient, who has been prepared for a surgical procedure. The sterile field may include the scrubbed team members, who are properly attired, and all furniture and fixtures in the area.

Referring now to FIG. 3 , a hub 106 is depicted in communication with a visualization system 108, a robotic system 110, and a handheld intelligent surgical instrument 112. The huh 106 includes a huh display 135, an imaging module 138, a generator module 140, a communication module 130, a processor module 132, a storage array 134, and an operating-room mapping module 133. In certain aspects, as illustrated in FIG. 3 , the hub 106 further includes a smoke evacuation module 126 and/or a suction/irrigation module 128. During a surgical procedure, energy application to tissue, for sealing and/or cutting, is generally associated with smoke evacuation, suction of excess fluid, and/or irrigation of the tissue. Fluid, power, and/or data lines from different sources are often entangled during the surgical procedure. Valuable time can be lost addressing this issue during a surgical procedure. Detangling the lines may necessitate disconnecting the lines from their respective modules, which may require resetting the modules. The huh modular enclosure 136 offers a unified environment for managing the power, data, and fluid lines, which reduces the frequency of entanglement between such lines. Aspects of the present disclosure present a surgical hub for use in a surgical procedure that involves energy application to tissue at a surgical site. The surgical hub includes a hub enclosure and a combo generator module slidably receivable in a docking station of the hub enclosure. The docking station includes data and power contacts. The combo generator module includes two or more of an ultrasonic energy generator component, a bipolar RF energy generator component, and a monopolar RF energy generator component that are housed in a single unit. In one aspect, the combo generator module also includes a smoke evacuation component, at least one energy delivery cable for connecting the combo generator module to a surgical instrument, at least one smoke evacuation component configured to evacuate smoke, fluid, and/or particulates generated by the application of therapeutic energy to the tissue, and a fluid line extending from the remote surgical site to the smoke evacuation component. In one aspect, the fluid line is a first fluid line and a second fluid line extends from the remote surgical site to a suction and irrigation module slidably received in the hub enclosure. In one aspect, the hub enclosure comprises a fluid interface. Certain surgical procedures may require the application of more than one energy type to the tissue. One energy type may be more beneficial for cutting the tissue, while another different energy type may be more beneficial for sealing the tissue. For example, a bipolar generator can be used to seal the tissue while an ultrasonic generator can be used to cut the sealed tissue. Aspects of the present disclosure present a solution where a huh modular enclosure 136 is configured to accommodate different generators, and facilitate an interactive communication therebetween. One of the advantages of the hub modular enclosure 136 is enabling the quick removal and/or replacement of various modules. Aspects of the present disclosure present a modular surgical enclosure for use in a surgical procedure that involves energy application to tissue. The modular surgical enclosure includes a first energy-generator module, configured to generate a first energy for application to the tissue, and a first docking station comprising a first docking port that includes first data and power contacts, wherein the first energy-generator module is slidably movable into an electrical engagement with the power and data contacts and wherein the first energy-generator module is slidably movable out of the electrical engagement with the first power and data contacts. Further to the above, the modular surgical enclosure also includes a second energy-generator module configured to generate a second energy, different than the first energy, for application to the tissue, and a second docking station comprising a second docking port that includes second data and power contacts, wherein the second energy-generator module is slidably movable into an electrical engagement with the power and data contacts, and wherein the second energy-generator module is slidably movable out of the electrical engagement with the second power and data contacts. In addition, the modular surgical enclosure also includes a communication bus between the first docking port and the second docking port, configured to facilitate communication between the first energy-generator module and the second energy-generator module. Referring to FIG. 3 , aspects of the present disclosure are presented for a hub modular enclosure 136 that allows the modular integration of a generator module 140, a smoke evacuation module 126, and a suction/irrigation module 128. The hub modular enclosure 136 further facilitates interactive communication between the modules 140, 126, 128. The generator module 140 can be a generator module with integrated monopolar, bipolar, and ultrasonic components supported in a single housing unit slidably insertable into the hub modular enclosure 136. The generator module 140 can be configured to connect to a monopolar device 142, a bipolar device 144, and an ultrasonic device 146. Alternatively, the generator module 140 may comprise a series of monopolar, bipolar, and/or ultrasonic generator modules that interact through the huh modular enclosure 136. The huh modular enclosure 136 can be configured to facilitate the insertion of multiple generators and interactive communication between the generators docked into the hub modular enclosure 136 so that the generators would act as a single generator.

FIG. 4 illustrates a surgical data network 201 comprising a modular communication hub 203 configured to connect modular devices located in one or more operating theaters of a healthcare facility, or any room in a healthcare facility specially equipped for surgical operations, to a cloud-based system (e.g., the cloud 204 that may include a remote server 213 coupled to a storage device 205). In one aspect, the modular communication hub 203 comprises a network hub 207 and/or a network switch 209 in communication with a network router. The modular communication hub 203 also can be coupled to a local computer system 210 to provide local computer processing and data manipulation. The surgical data network 201 may be configured as passive, intelligent, or switching. A passive surgical data network serves as a conduit for the data, enabling it to go from one device (or segment) to another and to the cloud computing resources. An intelligent surgical data network includes additional features to enable the traffic passing through the surgical data network to be monitored and to configure each port in the network huh 207 or network switch 209. An intelligent surgical data network may be referred to as a manageable hub or switch. A switching hub reads the destination address of each packet and then forwards the packet to the correct port.

Modular devices 1 a-1 n located in the operating theater may be coupled to the modular communication hub 203. The network hub 207 and/or the network switch 209 may be coupled to a network router 211 to connect the devices 1 a-1 n to the cloud 204 or the local computer system 210. Data associated with the devices 1 a-1 n may be transferred to cloud-based computers via the router for remote data processing and manipulation. Data associated with the devices 1 a-1 n may also be transferred to the local computer system 210 for local data processing and manipulation. Modular devices 2 a-2 m located in the same operating theater also may be coupled to a network switch 209. The network switch 209 may be coupled to the network hub 207 and/or the network router 211 to connect to the devices 2 a-2 m to the cloud 204. Data associated with the devices 2 a-2 n may be transferred to the cloud 204 via the network router 211 for data processing and manipulation. Data associated with the devices 2 a-2 m may also be transferred to the local computer system 210 for local data processing and manipulation.

It will be appreciated that the surgical data network 201 may be expanded by interconnecting multiple network hubs 207 and/or multiple network switches 209 with multiple network routers 211. The modular communication hub 203 may be contained in a modular control tower configured to receive multiple devices 1 a-1 n/2 a-2 m. The local computer system 210 also may be contained in a modular control tower. The modular communication hub 203 is connected to a display 212 to display images obtained by some of the devices 1 a-1 n/2 a-2 m, for example during surgical procedures. In various aspects, the devices 1 a-1 n/2 a-2 m may include, for example, various modules such as an imaging module 138 coupled to an endoscope, a generator module 140 coupled to an energy-based surgical device, a smoke evacuation module 126, a suction/irrigation module 128, a communication module 130, a processor module 132, a storage array 134, a surgical device coupled to a display, and/or a non-contact sensor module, among other modular devices that may be connected to the modular communication hub 203 of the surgical data network 201.

In one aspect, the surgical data network 201 may comprise a combination of network hub(s), network switch(es), and network router(s) connecting the devices 1 a-1 n/2 a-2 m to the cloud. Any one of or all of the devices 1 a-1 n/2 a-2 m coupled to the network hub or network switch may collect data in real time and transfer the data to cloud computers for data processing and manipulation. It will be appreciated that cloud computing relies on sharing computing resources rather than having local servers or personal devices to handle software applications. The word “cloud” may be used as a metaphor for “the Internet,” although the term is not limited as such. Accordingly, the term “cloud computing” may be used herein to refer to “a type of Internet-based computing,” where different services—such as servers, storage, and applications—are delivered to the modular communication hub 203 and/or computer system 210 located in the surgical theater (e.g., a fixed, mobile, temporary, or field operating room or space) and to devices connected to the modular communication hub 203 and/or computer system 210 through the Internet. The cloud infrastructure may be maintained by a cloud service provider. In this context, the cloud service provider may be the entity that coordinates the usage and control of the devices 1 a-1 n/2 a-2 m located in one or more operating theaters. The cloud computing services can perform a large number of calculations based on the data gathered by smart surgical instruments, robots, and other computerized devices located in the operating theater. The hub hardware enables multiple devices or connections to be connected to a computer that communicates with the cloud computing resources and storage.

Applying cloud computer data processing techniques on the data collected by the devices 1 a-1 n/2 a-2 m, the surgical data network can provide improved surgical outcomes, reduced costs, and improved patient satisfaction. At least some of the devices 1 a-1 n/2 a-2 m may be employed to view tissue states to assess leaks or perfusion of sealed tissue after a tissue sealing and cutting procedure. At least some of the devices 1 a-1 n/2 a-2 m may be employed to identify pathology, such as the effects of diseases, using the cloud-based computing to examine data including images of samples of body tissue for diagnostic purposes. This may include localization and margin confirmation of tissue and phenotypes. At least some of the devices 1 a-1 n/2 a-2 m may be employed to identify anatomical structures of the body using a variety of sensors integrated with imaging devices and techniques such as overlaying images captured by multiple imaging devices. The data gathered by the devices 1 a-1 n/2 a-2 m, including image data, may be transferred to the cloud 204 or the local computer system 210 or both for data processing and manipulation including image processing and manipulation. The data may be analyzed to improve surgical procedure outcomes by determining if further treatment, such as the application of endoscopic intervention, emerging technologies, a targeted radiation, targeted intervention, and precise robotics to tissue-specific sites and conditions, may be pursued. Such data analysis may further employ outcome analytics processing, and using standardized approaches may provide beneficial feedback to either confirm surgical treatments and the behavior of the surgeon or suggest modifications to surgical treatments and the behavior of the surgeon.

The operating theater devices 1 a-1 n may be connected to the modular communication hub 203 over a wired channel or a wireless channel depending on the configuration of the devices 1 a-1 n to a network hub. The network hub 207 may be implemented, in one aspect, as a local network broadcast device that works on the physical layer of the Open System Interconnection (OSI) model. The network hub may provide connectivity to the devices 1 a-1 n located in the same operating theater network. The network hub 207 may collect data in the form of packets and sends them to the router in half duplex mode. The network hub 207 may not store any media access control/Internet Protocol (MAC/IP) to transfer the device data. Only one of the devices 1 a-1 n can send data at a time through the network hub 207. The network hub 207 may not have routing tables or intelligence regarding where to send information and broadcasts all network data across each connection and to a remote server 213 (FIG. 4 ) over the cloud 204. The network hub 207 can detect basic network errors such as collisions, but having all information broadcast to multiple ports can be a security risk and cause bottlenecks.

The operating theater devices 2 a-2 m may be connected to a network switch 209 over a wired channel or a wireless channel. The network switch 209 works in the data link layer of the OSI model. The network switch 209 may be a multicast device for connecting the devices 2 a-2 m located in the same operating theater to the network. The network switch 209 may send data in the form of frames to the network router 211 and works in full duplex mode. Multiple devices 2 a-2 m can send data at the same time through the network switch 209. The network switch 209 stores and uses MAC addresses of the devices 2 a-2 m to transfer data.

The network hub 207 and/or the network switch 209 may be coupled to the network router 211 for connection to the cloud 204. The network router 211 works in the network layer of the OSI model. The network router 211 creates a route for transmitting data packets received from the network hub 207 and/or network switch 211 to cloud-based computer resources for further processing and manipulation of the data collected by any one of or all the devices 1 a-1 n/2 a-2 m. The network router 211 may be employed to connect two or more different networks located in different locations, such as, for example, different operating theaters of the same healthcare facility or different networks located in different operating theaters of different healthcare facilities. The network router 211 may send data in the form of packets to the cloud 204 and works in full duplex mode. Multiple devices can send data at the same time. The network router 211 uses IP addresses to transfer data.

In an example, the network hub 207 may be implemented as a USB hub, which allows multiple USB devices to be connected to a host computer. The USB hub may expand a single USB port into several tiers so that there are more ports available to connect devices to the host system computer. The network hub 207 may include wired or wireless capabilities to receive information over a wired channel or a wireless channel. In one aspect, a wireless USB short-range, high-bandwidth wireless radio communication protocol may be employed for communication between the devices 1 a-1 n and devices 2 a-2 m located in the operating theater.

In examples, the operating theater devices 1 a-1 n/2 a-2 m may communicate to the modular communication hub 203 via Bluetooth wireless technology standard for exchanging data over short distances (using short-wavelength UHF radio waves in the ISM band from 2.4 to 2.485 GHz) from fixed and mobile devices and building personal area networks (PANs). The the operating theater devices 1 a-1 n/2 a-2 m may communicate to the modular communication hub 203 via a number of wireless or wired communication standards or protocols, including but not limited to Wi-Fi (IEEE 802.11 family), WiMAX (IEEE 802.16 family), IEEE 802.20, new radio (NR), long-tenn evolution (LTE), and Ev-DO, HSPA+, HSDPA+, HSUPA+, EDGE, GSM, GPRS, CDMA, TDMA, DECT, and Ethernet derivatives thereof, as well as any other wireless and wired protocols that are designated as 3G, 4G, 5G, and beyond. The computing module may include a plurality of communication modules. For instance, a first communication module may be dedicated to shorter-range wireless communications such as Wi-Fi and Bluetooth, and a second communication module may be dedicated to longer-range wireless communications such as GPS, EDGE, GPRS, CDMA, WiMAX, LTE, Ev-DO, and others.

The modular communication hub 203 may serve as a central connection for one or all of the operating theater devices 1 a-1 n/2 a-2 m and may handle a data type known as francs. Frames may carry the data generated by the devices 1 a-1 n/2 a-2 m. When a frame is received by the modular communication hub 203, it is amplified and transmitted to the network router 211, which transfers the data to the cloud computing resources by using a number of wireless or wired communication standards or protocols, as described herein.

The modular communication hub 203 can be used as a standalone device or be connected to compatible network hubs and network switches to form a larger network. The modular communication hub 203 can be generally easy to install, configure, and maintain, making it a good option for networking the operating theater devices 1 a-In/2 a-2 m.

FIG. 5 illustrates a computer-implemented interactive surgical system 200. The computer-implemented interactive surgical system 200 is similar in many respects to the computer-implemented interactive surgical system 100. For example, the computer-implemented interactive surgical system 200 includes one or more surgical systems 202, which are similar in many respects to the surgical systems 102. Each surgical system 202 includes at least one surgical hub 206 in communication with a cloud 204 that may include a remote server 213. In one aspect, the computer-implemented interactive surgical system 200 comprises a modular control tower 236 connected to multiple operating theater devices such as, for example, intelligent surgical instruments, robots, and other computerized devices located in the operating theater. As shown in FIG. 6 , the modular control tower 236 comprises a modular communication hub 203 coupled to a computer system 210.

As illustrated in the example of FIG. 5 , the modular control tower 236 may be coupled to an imaging module 238 that may be coupled to an endoscope 239, a generator module 240 that may be coupled to an energy device 241, a smoke evacuator module 226, a suction/irrigation module 228, a communication module 230, a processor module 232, a storage array 234, a smart device/instrument 235 optionally coupled to a display 237, and a non-contact sensor module 242. The operating theater devices may be coupled to cloud computing resources and data storage via the modular control tower 236. A robot hub 222 also may be connected to the modular control tower 236 and to the cloud computing resources. The devices/instruments 235, visualization systems 208, among others, may be coupled to the modular control tower 236 via wired or wireless communication standards or protocols, as described herein. The modular control tower 236 may be coupled to a hub display 215 (e.g., monitor, screen) to display and overlay images received from the imaging module, device/instrument display, and/or other visualization systems 208. The hub display also may display data received from devices connected to the modular control tower in conjunction with images and overlaid images.

FIG. 6 illustrates a surgical hub 206 comprising a plurality of modules coupled to the modular control tower 236. The modular control tower 236 may comprise a modular communication hub 203, e.g., a network connectivity device, and a computer system 210 to provide local processing, visualization, and imaging, for example. As shown in FIG. 6 , the modular communication hub 203 may be connected in a tiered configuration to expand the number of modules (e.g., devices) that may be connected to the modular communication hub 203 and transfer data associated with the modules to the computer system 210, cloud computing resources, or both. As shown in FIG. 6 , each of the network hubs/switches in the modular communication hub 203 may include three downstream ports and one upstream port. The upstream network hub/switch may be connected to a processor to provide a communication connection to the cloud computing resources and a local display 217. Communication to the cloud 204 may be made either through a wired or a wireless communication channel.

The surgical hub 206 may employ a non-contact sensor module 242 to measure the dimensions of the operating theater and generate a map of the surgical theater using either ultrasonic or laser-type non-contact measurement devices. An ultrasound-based non-contact sensor module may scan the operating theater by transmitting a burst of ultrasound and receiving the echo when it bounces off the perimeter walls of an operating theater as described under the heading “Surgical Hub Spatial Awareness Within an Operating Room” in .S. Patent Application Publication No. US 2019-0200844 A1 (U.S. patent application Ser. No. 16/209,385), titled METHOD OF HUB COMMUNICATION, PROCESSING, STORAGE AND DISPLAY, filed Dec. 4, 2018, which is herein incorporated by reference in its entirety, in which the sensor module is configured to determine the size of the operating theater and to adjust Bluetooth-pairing distance limits. A laser-based non-contact sensor module may scan the operating theater by transmitting laser light pulses, receiving laser light pulses that bounce off the perimeter walls of the operating theater, and comparing the phase of the transmitted pulse to the received pulse to determine the size of the operating theater and to adjust Bluetooth pairing distance limits, for example.

The computer system 210 may comprise a processor 244 and a network interface 245. The processor 244 can be coupled to a communication module 247, storage 248, memory 249, non-volatile memory 250, and input/output interface 251 via a system bus. The system bus can be any of several types of bus structure(s) including the memory bus or memory controller, a peripheral bus or external bus, and/or a local bus using any variety of available bus architectures including, but not limited to, 9-bit bus, Industrial Standard Architecture (ISA), Micro-Charmel Architecture (MSA), Extended ISA (EISA), Intelligent Drive Electronics (IDE), VESA Local Bus (VLB), Peripheral Component Interconnect (PCI), USB, Advanced Graphics Port (AGP), Personal Computer Memory Card International Association bus (PCMCIA), Small Computer Systems Interface (SCSI), or any other proprietary bus.

The processor 244 may be any single-core or multicore processor such as those known under the trade name ARM Cortex by Texas Instruments. In one aspect, the processor may be an LM4F230H5QR ARM Cortex-M4F Processor Core, available from Texas Instruments, for example, comprising an on-chip memory of 256 KB single-cycle flash memory, or other non-volatile memory, up to 40 MHz, a prefetch buffer to improve performance above 40 MI Iz, a 32 KB single-cycle serial random access memory (SRAM), an internal read-only memory (ROM) loaded with StellarisWare® software, a 2 KB electrically erasable programmable read-only memory (FEPROM), and/or one or more pulse width modulation (PWM) modules, one or more quadrature encoder inputs (QEI) analogs, one or more 12-bit analog-to-digital converters (ADCs) with 12 analog input channels, details of which are available for the product datasheet.

In one aspect, the processor 244 may comprise a safety controller comprising two controller-based families such as TMS570 and RM4x, known under the trade name Hercules ARM Cortex R4, also by Texas Instruments. The safety controller may be configured specifically for IEC 61508 and ISO 26262 safety critical applications, among others, to provide advanced integrated safety features while delivering scalable performance, connectivity, and memory options.

The system memory may include volatile memory and non-volatile memory. The basic input/output system (BIOS), containing the basic routines to transfer information between elements within the computer system, such as during start-up, is stored in non-volatile memory. For example, the non-volatile memory can include ROM, programmable ROM (PROM), electrically programmable ROM (EPROM), EEPROM, or flash memory. Volatile memory includes random-access memory (RAM), which acts as external cache memory. Moreover, RAM is available in many forms such as SRAM, dynamic RAM (DRAM), synchronous DRAM (SDRAM), double data rate SDRAM (DDR SDRAM), enhanced SDRAM (ESDRAM), Synchlink DRAM (SLDRAM), and direct Rambus RAM (DRRAM).

The computer system 210 also may include removable/non-removable, volatile/non-volatile computer storage media, such as for example disk storage. The disk storage can include, but is not limited to, devices like a magnetic disk drive, floppy disk drive, tape drive, Jaz drive, Zip drive, LS-60 drive, flash memory card, or memory stick. In addition, the disk storage can include storage media separately or in combination with other storage media including, but not limited to, an optical disc drive such as a compact disc ROM device (CD-ROM), compact disc recordable drive (CD-R Drive), compact disc rewritable drive (CD-RW Drive), or a digital versatile disc ROM drive (DVD-ROM). To facilitate the connection of the disk storage devices to the system bus, a removable or non-removable interface may be employed.

It is to be appreciated that the computer system 210 may include software that acts as an intermediary between users and the basic computer resources described in a suitable operating environment. Such software may include an operating system. The operating system, which can be stored on the disk storage, may act to control and allocate resources of the computer system. System applications may take advantage of the management of resources by the operating system through program modules and program data stored either in the system memory or on the disk storage. It is to be appreciated that various components described herein can be implemented with various operating systems or combinations of operating systems.

A user may enter commands or information into the computer system 210 through input device(s) coupled to the I/O interface 251. The input devices may include, but are not limited to, a pointing device such as a mouse, trackball, stylus, touch pad, keyboard, microphone, joystick, game pad, satellite dish, scanner, TV tuner card, digital camera, digital video camera, web camera, and the like. These and other input devices connect to the processor through the system bus via interface port(s). The interface port(s) include, for example, a serial port, a parallel port, a game port, and a USB. The output device(s) use some of the same types of ports as input device(s). Thus, for example, a USB port may be used to provide input to the computer system and to output information from the computer system to an output device. An output adapter may be provided to illustrate that there can be some output devices like monitors, displays, speakers, and printers, among other output devices that may require special adapters. The output adapters may include, by way of illustration and not limitation, video and sound cards that provide a means of connection between the output device and the system bus. It should be noted that other devices and/or systems of devices, such as remote computer(s), may provide both input and output capabilities.

The computer system 210 can operate in a networked environment using logical connections to one or more remote computers, such as cloud computer(s), or local computers. The remote cloud computer(s) can be a personal computer, server, router, network PC, workstation, microprocessor-based appliance, peer device, or other common network node, and the like, and typically includes many or all of the elements described relative to the computer system. For purposes of brevity, only a memory storage device is illustrated with the remote computer(s). The remote computer(s) may be logically connected to the computer system through a network interface and then physically connected via a communication connection. The network interface may encompass communication networks such as local area networks (LANs) and wide area networks (WANs). LAN technologies may include Fiber Distributed Data Interface (MDT), Copper Distributed Data Interface (CDDI), Ethernet/IEEE 802.3, Token Ring/IEEE 802.5 and the like. WAN technologies may include, but are not limited to, point-to-point links, circuit-switching networks like Integrated Services Digital Networks (ISDN) and variations thereon, packet-switching networks, and Digital Subscriber Lines (DSL).

In various aspects, the computer system 210 of FIG. 6 , the imaging module 238 and/or visualization system 208, and/or the processor module 232 of FIGS. 5-6 , may comprise an image processor, image-processing engine, media processor, or any specialized digital signal processor (DSP) used for the processing of digital images. The image processor may employ parallel computing with single instruction, multiple data (SIMD) or multiple instruction, multiple data (MIMD) technologies to increase speed and efficiency. The digital image-processing engine can perform a range of tasks. The image processor may be a system on a chip with multicore processor architecture.

The communication connection(s) may refer to the hardware/software employed to connect the network interface to the bus. While the communication connection is shown for illustrative clarity inside the computer system, it can also be external to the computer system 210. The hardware/software necessary for connection to the network interface may include, for illustrative purposes only, internal and external technologies such as modems, including regular telephone-grade modems, cable modems, and DSL modems, ISDN adapters, and Ethernet cards.

FIG. 7 illustrates a logic diagram of a control system 470 of a surgical instrument or tool in accordance with one or more aspects of the present disclosure. The system 470 may comprise a control circuit. The control circuit may include a microcontroller 461 comprising a processor 462 and a memory 468. One or more of sensors 472, 474, 476, for example, provide real-time feedback to the processor 462. A motor 482, driven by a motor driver 492, operably couples a longitudinally movable displacement member to drive the I-beam knife element. A tracking system 480 may be configured to determine the position of the longitudinally movable displacement member. The position information may be provided to the processor 462, which can be programmed or configured to determine the position of the longitudinally movable drive member as well as the position of a firing member, firing bar, and I-beam knife element. Additional motors may be provided at the tool driver interface to control I-beam firing, closure tube travel, shaft rotation, and articulation. A display 473 may display a variety of operating conditions of the instruments and may include touch screen functionality for data input. Information displayed on the display 473 may be overlaid with images acquired via endoscopic imaging modules.

In one aspect, the microcontroller 461 may be any single-core or multicore processor such as those known under the trade name ARM Cortex by Texas Instruments. In one aspect, the main microcontroller 461 may be an LM4F230H5QR ARM Cortex-M4F Processor Core, available from Texas Instruments, for example, comprising an on-chip memory of 256 KB single-cycle flash memory, or other non-volatile memory, up to 40 MHz, a prefetch buffer to improve performance above 40 MHz, a 32 KB single-cycle SRAM, and internal ROM loaded with StellarisWare® software, a 2 KB EEPROM, one or more PWM modules, one or more QEI analogs, and/or one or more 12-bit ADCs with 12 analog input channels, details of which are available for the product datasheet.

In one aspect, the microcontroller 461 may comprise a safety controller comprising two controller-based families such as TMS570 and RM4x, known under the trade name Hercules ARM Cortex R4, also by Texas Instruments. The safety controller may be configured specifically for IEC 61508 and ISO 26262 safety critical applications, among others, to provide advanced integrated safety features while delivering scalable performance, connectivity, and memory options.

The microcontroller 461 may be programmed to perform various functions such as precise control over the speed and position of the knife and articulation systems. In one aspect, the microcontroller 461 may include a processor 462 and a memory 468. The electric motor 482 may be a brushed direct current (DC) motor with a gearbox and mechanical links to an articulation or knife system. In one aspect, a motor driver 492 may be an A3941 available from Allegro Microsystems, Inc. Other motor drivers may be readily substituted for use in the tracking system 480 comprising an absolute positioning system. A detailed description of an absolute positioning system is described in U.S. Patent Application Publication No. 2017/0296213, titled SYSTEMS AND METHODS FOR CONTROLLING A SURGICAL STAPLING AND CUTTING INSTRUMENT, which published on Oct. 19, 2017, which is herein incorporated by reference in its entirety.

The microcontroller 461 may be programmed to provide precise control over the speed and position of displacement members and articulation systems. The microcontroller 461 may be configured to compute a response in the software of the microcontroller 461. The computed response may be compared to a measured response of the actual system to obtain an “observed” response, which is used for actual feedback decisions. The observed response may be a favorable, tuned value that balances the smooth, continuous nature of the simulated response with the measured response, which can detect outside influences on the system.

In some examples, the motor 482 may be controlled by the motor driver 492 and can be employed by the firing system of the surgical instrument or tool. In various forms, the motor 482 may be a brushed DC driving motor having a maximum rotational speed of approximately 25,000 RPM. In some examples, the motor 482 may include a brushless motor, a cordless motor, a synchronous motor, a stepper motor, or any other suitable electric motor. The motor driver 492 may comprise an H-bridge driver comprising field-effect transistors (FETs), for example. The motor 482 can be powered by a power assembly releasably mounted to the handle assembly or tool housing for supplying control power to the surgical instrument or tool. The power assembly may comprise a battery which may include a number of battery cells connected in series that can be used as the power source to power the surgical instrument or tool. In certain circumstances, the battery cells of the power assembly may be replaceable and/or rechargeable. In at least one example, the battery cells can be lithium-ion batteries which can be couplable to and separable from the power assembly.

The motor driver 492 may be an A3941 available from Allegro Microsystems, Inc. The A3941 492 may be a full-bridge controller for use with external N-channel power metal-oxide semiconductor field-effect transistors (MOSFETs) specifically designed for inductive loads, such as brush DC motors. The driver 492 may comprise a unique charge pump regulator that can provide full (>10 V) gate drive for battery voltages down to 7 V and can allow the A3941 to operate with a reduced gate drive, down to 5.5 V. A bootstrap capacitor may be employed to provide the above battery supply voltage required for N-channel MOSFETs. An internal charge pump for the high-side drive may allow DC (100% duty cycle) operation. The full bridge can be driven in fast or slow decay modes using diode or synchronous rectification. In the slow decay mode, current recirculation can be through the high-side or the lowside FETs. The power FETs may be protected from shoot-through by resistor-adjustable dead tune. Integrated diagnostics provide indications of undervoltage, overtemperature, and power bridge faults and can be configured to protect the power MOSFETs under most short circuit conditions. Other motor drivers may be readily substituted for use in the tracking system 480 comprising an absolute positioning system.

The tracking system 480 may comprise a controlled motor drive circuit arrangement comprising a position sensor 472 according to one aspect of this disclosure. The position sensor 472 for an absolute positioning system may provide a unique position signal corresponding to the location of a displacement member. In some examples, the displacement member may represent a longitudinally movable drive member comprising a rack of drive teeth for meshing engagement with a corresponding drive gear of a gear reducer assembly. In some examples, the displacement member may represent the firing member, which could be adapted and configured to include a rack of drive teeth. In some examples, the displacement member may represent a firing bar or the I-beam, each of which can be adapted and configured to include a rack of drive teeth. Accordingly, as used herein, the term displacement member can be used generically to refer to any movable member of the surgical instrument or tool such as the drive member, the firing member, the firing bar, the I-beam, or any element that can be displaced. In one aspect, the longitudinally movable drive member can be coupled to the firing member, the firing bar, and the I-beam. Accordingly, the absolute positioning system can, in effect, track the linear displacement of the I-beam by tracking the linear displacement of the longitudinally movable drive member. In various aspects, the displacement member may be coupled to any position sensor 472 suitable for measuring linear displacement. Thus, the longitudinally movable drive member, the firing member, the firing bar, or the I-beam, or combinations thereof, may be coupled to any suitable linear displacement sensor. Linear displacement sensors may include contact or non-contact displacement sensors. Linear displacement sensors may comprise linear variable differential transformers (LVDT), differential variable reluctance transducers (DVRT), a slide potentiometer, a magnetic sensing system comprising a movable magnet and a series of linearly arranged Hall effect sensors, a magnetic sensing system comprising a fixed magnet and a series of movable, linearly arranged Hall effect sensors, an optical sensing system comprising a movable light source and a series of linearly arranged photo diodes or photo detectors, an optical sensing system comprising a fixed light source and a series of movable linearly, arranged photo diodes or photo detectors, or any combination thereof.

The electric motor 482 can include a rotatable shaft that operably interfaces with a gear assembly that is mounted in meshing engagement with a set, or rack, of drive teeth on the displacement member. A sensor element may be operably coupled to a gear assembly such that a single revolution of the position sensor 472 element corresponds to some linear longitudinal translation of the displacement member. An arrangement of gearing and sensors can be connected to the linear actuator, via a rack and pinion arrangement, or a rotary actuator, via a spur gear or other connection. A power source may supplie power to the absolute positioning system and an output indicator may display the output of the absolute positioning system. The displacement member may represent the longitudinally movable drive member comprising a rack of drive teeth formed thereon for meshing engagement with a corresponding drive gear of the gear reducer assembly. The displacement member may represent the longitudinally movable firing member, firing bar, I-beam, or combinations thereof.

A single revolution of the sensor element associated with the position sensor 472 may be equivalent to a longitudinal linear displacement d1 of the of the displacement member, where d1 is the longitudinal linear distance that the displacement member moves from point “a” to point “b” after a single revolution of the sensor element coupled to the displacement member. The sensor arrangement may be connected via a gear reduction that results in the position sensor 472 completing one or more revolutions for the full stroke of the displacement member. The position sensor 472 may complete multiple revolutions for the full stroke of the displacement member.

A series of switches, where n is an integer greater than one, may be employed alone or in combination with a gear reduction to provide a unique position signal for more than one revolution of the position sensor 472. The state of the switches may be fed back to the microcontroller 461 that applies logic to determine a unique position signal corresponding to the longitudinal linear displacement d1+d2+ . . . dn of the displacement member. The output of the position sensor 472 is provided to the microcontroller 461. The position sensor 472 of the sensor arrangement may comprise a magnetic sensor, an analog rotary sensor like a potentiometer, or an array of analog Hall-effect elements, which output a unique combination of position signals or values.

The position sensor 472 may comprise any number of magnetic sensing elements, such as, for example, magnetic sensors classified according to whether they measure the total magnetic field or the vector components of the magnetic field. The techniques used to produce both types of magnetic sensors may encompass many aspects of physics and electronics. The technologies used for magnetic field sensing may include search coil, fluxgate, optically pumped, nuclear precession, SQUID, Hall-effect, anisotropic magnetoresistance, giant magnetoresistance, magnetic tunnel junctions, giant magnetoimpedance, magnetostrictive/piezoelectric composites, magnetodiode, magnetotransistor, fiber-optic, magneto-optic, and microelectromechanical systems-based magnetic sensors, among others.

In one aspect, the position sensor 472 for the tracking system 480 comprising an absolute positioning system may comprise a magnetic rotary absolute positioning system. The position sensor 472 may be implemented as an AS5055EQF1 single-chip magnetic rotary position sensor available from Austria Microsystems, AG. The position sensor 472 is interfaced with the microcontroller 461 to provide an absolute positioning system. The position sensor 472 may be a low-voltage and low-power component and includes four Hall-effect elements in an area of the position sensor 472 that may be located above a magnet. A high-resolution ADC and a smart power management controller may also be provided on the chip. A coordinate rotation digital computer (CORDIC) processor, also known as the digit-by-digit method and Volder's algorithm, may be provided to implement a simple and efficient algorithm to calculate hyperbolic and trigonometric functions that require only addition, subtraction, bitshift, and table lookup operations. The angle position, alarm bits, and magnetic field information may be transmitted over a standard serial communication interface, such as a serial peripheral interface (SPI) interface, to the microcontroller 461. The position sensor 472 may provide 12 or 14 bits of resolution. The position sensor 472 may be an AS5055 chip provided in a small QFN 16-pin 4×4×0.85 mm package.

The tracking system 480 comprising an absolute positioning system may comprise and/or be programmed to implement a feedback controller, such as a PID, state feedback, and adaptive controller. A power source converts the signal from the feedback controller into a physical input to the system: in this case the voltage. Other examples include a PWM of the voltage, current, and force. Other sensor(s) may be provided to measure physical parameters of the physical system in addition to the position measured by the position sensor 472. In some aspects, the other sensor(s) can include sensor arrangements such as those described in U.S. Pat. No. 9,345,481, titled STAPLE CARTRIDGE TISSUE THICKNESS SENSOR SYSTEM, which issued on May 24, 2016, which is herein incorporated by reference in its entirety; U.S. Patent Application Publication No. 2014/0263552, titled STAPLE CARTRIDGE TISSUE THICKNESS SENSOR SYSTEM, which published on Sep. 18, 2014, which is herein incorporated by reference in its entirety; and U.S. patent application Ser. No. 15/628,175, titled TECHNIQUES FOR ADAPTIVE CONTROL OF MOTOR VELOCITY OF A SURGICAL STAPLING AND CUTTING INSTRUMENT, filed Jun. 20, 2017, which is herein incorporated by reference in its entirety. In a digital signal processing system, an absolute positioning system is coupled to a digital data acquisition system where the output of the absolute positioning system will have a finite resolution and sampling frequency. The absolute positioning system may comprise a compare-and-combine circuit to combine a computed response with a measured response using algorithms, such as a weighted average and a theoretical control loop, that drive the computed response towards the measured response. The computed response of the physical system may take into account properties like mass, inertial, viscous friction, inductance resistance, etc., to predict what the states and outputs of the physical system will be by knowing the input.

The absolute positioning system may provide an absolute position of the displacement member upon power-up of the instrument, without retracting or advancing the displacement member to a reset (zero or home) position as may be required with conventional rotary encoders that merely count the number of steps forwards or backwards that the motor 482 has taken to infer the position of a device actuator, drive bar, knife, or the like.

A sensor 474, such as, for example, a strain gauge or a micro-strain gauge, may be configured to measure one or more parameters of the end effector, such as, for example, the amplitude of the strain exerted on the anvil during a clamping operation, which can be indicative of the closure forces applied to the anvil. The measured strain may be converted to a digital signal and provided to the processor 462. Alternatively, or in addition to the sensor 474, a sensor 476, such as, for example, a load sensor, can measure the closure force applied by the closure drive system to the anvil. The sensor 476, such as, for example, a load sensor, can measure the firing force applied to an I-beam in a firing stroke of the surgical instrument or tool. The I-beam is configured to engage a wedge sled, which is configured to upwardly cam staple drivers to force out staples into deforming contact with an anvil. The I-beam also may include a sharpened cutting edge that can be used to sever tissue as the I-beam is advanced distally by the firing bar. Alternatively, a current sensor 478 can be employed to measure the current drawn by the motor 482. The force required to advance the firing member can correspond to the current drawn by the motor 482, for example. The measured force may be converted to a digital signal and provided to the processor 462.

In one form, the strain gauge sensor 474 can be used to measure the force applied to the tissue by the end effector. A strain gauge can be coupled to the end effector to measure the force on the tissue being treated by the end effector. A system for measuring forces applied to the tissue grasped by the end effector may comprise a strain gauge sensor 474, such as, for example, a micro-strain gauge, that can be configured to measure one or more parameters of the end effector, for example. In one aspect, the strain gauge sensor 474 can measure the amplitude or magnitude of the strain exerted on a jaw member of an end effector during a clamping operation, which can be indicative of the tissue compression. The measured strain can be converted to a digital signal and provided to a processor 462 of the microcontroller 461. A load sensor 476 can measure the force used to operate the knife element, for example, to cut the tissue captured between the anvil and the staple cartridge. A magnetic field sensor can be employed to measure the thickness of the captured tissue. The measurement of the magnetic field sensor also may be converted to a digital signal and provided to the processor 462.

The measurements of the tissue compression, the tissue thickness, and/or the force required to close the end effector on the tissue, as respectively measured by the sensors 474, 476, can be used by the microcontroller 461 to characterize the selected position of the firing member and/or the corresponding value of the speed of the firing member. In one instance, a memory 468 may store a technique, an equation, and/or a lookup table which can be employed by the microcontroller 461 in the assessment.

The control system 470 of the surgical instrument or tool also may comprise wired or wireless communication circuits to communicate with the modular communication hub 203 as shown in FIGS. 5 and 6 .

FIG. 8 illustrates a surgical instrument or tool comprising a plurality of motors which can be activated to perform various functions. In certain instances, a first motor can be activated to perform a first function, a second motor can be activated to perform a second function, a third motor can be activated to perform a third function, a fourth motor can be activated to perform a fourth function, and so on. In certain instances, the plurality of motors of robotic surgical instrument 600 can be individually activated to cause firing, closure, and/or articulation motions in the end effector. The firing, closure, and/or articulation motions can be transmitted to the end effector through a shaft assembly, for example.

In certain instances, the surgical instrument system or tool may include a firing motor 602. The firing motor 602 may be operably coupled to a firing motor drive assembly 604 which can be configured to transmit firing motions, generated by the motor 602 to the end effector, in particular to displace the I-beam element. In certain instances, the firing motions generated by the motor 602 may cause the staples to be deployed from the staple cartridge into tissue captured by the end effector and/or the cutting edge of the I-beam element to be advanced to cut the captured tissue, for example. The I-beam element may be retracted by reversing the direction of the motor 602.

In certain instances, the surgical instrument or tool may include a closure motor 603. The closure motor 603 may be operably coupled to a closure motor drive assembly 605 which can be configured to transmit closure motions, generated by the motor 603 to the end effector, in particular to displace a closure tube to close the anvil and compress tissue between the anvil and the staple cartridge. The closure motions may cause the end effector to transition from an open configuration to an approximated configuration to capture tissue, for example. The end effector may be transitioned to an open position by reversing the direction of the motor 603.

In certain instances, the surgical instrument or tool may include one or more articulation motors 606 a, 606 b, for example. The motors 606 a, 606 b may be operably coupled to respective articulation motor drive assemblies 608 a, 608 b, which can be configured to transmit articulation motions generated by the motors 606 a, 606 b to the end effector. Tn certain instances, the articulation motions may cause the end effector to articulate relative to the shaft, for example.

As described herein, the surgical instrument or tool may include a plurality of motors which may be configured to perform various independent functions. In certain instances, the plurality of motors of the surgical instrument or tool can be individually or separately activated to perform one or more functions while the other motors remain inactive. For example, the articulation motors 606 a, 606 b can be activated to cause the end effector to be articulated while the firing motor 602 remains inactive. Alternatively, the firing motor 602 can be activated to fire the plurality of staples, and/or to advance the cutting edge, while the articulation motor 606 remains inactive. Furthermore, the closure motor 603 may be activated simultaneously with the firing motor 602 to cause the closure tube and the I-beam element to advance distally as described in more detail hereinbelow.

In certain instances, the surgical instrument or tool may include a common control module 610 which can be employed with a plurality of motors of the surgical instrument or tool. In certain instances, the common control module 610 may accommodate one of the plurality of motors at a time. For example, the common control module 610 can be couplable to and separable from the plurality of motors of the robotic surgical instrument individually. In certain instances, a plurality of the motors of the surgical instrument or tool may share one or more common control modules such as the common control module 610. In certain instances, a plurality of motors of the surgical instrument or tool can be individually and selectively engaged with the common control module 610. In certain instances, the common control module 610 can be selectively switched from interfacing with one of a plurality of motors of the surgical instrument or tool to interfacing with another one of the plurality of motors of the surgical instrument or tool.

In at least one example, the common control module 610 can be selectively switched between operable engagement with the articulation motors 606 a, 606 b and operable engagement with either the firing motor 602 or the closure motor 603. In at least one example, as illustrated in FIG. 8 , a switch 614 can be moved or transitioned between a plurality of positions and/or states. In a first position 616, the switch 614 may electrically couple the common control module 610 to the firing motor 602; in a second position 617, the switch 614 may electrically couple the common control module 610 to the closure motor 603; in a third position 618 a, the switch 614 may electrically couple the common control module 610 to the first articulation motor 606 a; and in a fourth position 618 b, the switch 614 may electrically couple the common control module 610 to the second articulation motor 606 b, for example. In certain instances, separate common control modules 610 can be electrically coupled to the firing motor 602, the closure motor 603, and the articulations motor 606 a, 606 b at the same time. In certain instances, the switch 614 may be a mechanical switch, an electromechanical switch, a solid-state switch, or any suitable switching mechanism.

Each of the motors 602, 603, 606 a, 606 b may comprise a torque sensor to measure the output torque on the shaft of the motor. The force on an end effector may be sensed in any conventional manner, such as by force sensors on the outer sides of the jaws or by a torque sensor for the motor actuating the jaws.

In various instances, as illustrated in FIG. 8 , the common control module 610 may comprise a motor driver 626 which may comprise one or more H-Bridge FETs. The motor driver 626 may modulate the power transmitted from a power source 628 to a motor coupled to the common control module 610 based on input from a microcontroller 620 (the “controller”), for example. In certain instances, the microcontroller 620 can be employed to determine the current drawn by the motor, for example, while the motor is coupled to the common control module 610, as described herein.

In certain instances, the microcontroller 620 may include a microprocessor 622 (the “processor”) and one or more non-transitory computer-readable mediums or memory units 624 (the “memory”). In certain instances, the memory 624 may store various program instructions, which when executed may cause the processor 622 to perform a plurality of functions and/or calculations described herein. In certain instances, one or more of the memory units 624 may be coupled to the processor 622, for example.

In certain instances, the power source 628 can be employed to supply power to the microcontroller 620, for example. In certain instances, the power source 628 may comprise a battery (or “battery pack” or “power pack”), such as a lithium-ion battery, for example. In certain instances, the battery pack may be configured to be releasably mounted to a handle for supplying power to the surgical instrument 600. A number of battery cells connected in series may be used as the power source 628. In certain instances, the power source 628 may be replaceable and/or rechargeable, for example.

In various instances, the processor 622 may control the motor driver 626 to control the position, direction of rotation, and/or velocity of a motor that is coupled to the common control module 610. In certain instances, the processor 622 can signal the motor driver 626 to stop and/or disable a motor that is coupled to the common control module 610. It should be understood that the term “processor” as used herein includes any suitable microprocessor, microcontroller, or other basic computing device that incorporates the functions of a computer's central processing unit (CPU) on an integrated circuit or, at most, a few integrated circuits. The processor can be a multipurpose, programmable device that accepts digital data as input, processes it according to instructions stored in its memory, and provides results as output. It can be an example of sequential digital logic, as it may have internal memory. Processors may operate on numbers and symbols represented in the binary numeral system.

The processor 622 may be any single-core or multicore processor such as those known under the trade name ARM Cortex by Texas Instruments. In certain instances, the microcontroller 620 may be an LM 4F230H5QR, available from Texas Instruments, for example. In at least one example, the Texas Instruments LM4F230H5QR is an ARM Cortex-M4F Processor Core comprising an on-chip memory of 256 KB single-cycle flash memory, or other non-volatile memory, up to 40 MHz, a prefetch buffer to improve performance above 40 MHz, a 32 KB single-cycle SRAM, an internal ROM loaded with StellarisWare® software, a 2 KB EEPROM, one or more PWM modules, one or more QEI analogs, one or more 12-bit ADCs with 12 analog input channels, among other features that are readily available for the product datasheet. Other microcontrollers may be readily substituted for use with the module 4410. Accordingly, the present disclosure should not be limited in this context.

The memory 624 may include program instructions for controlling each of the motors of the surgical instrument 600 that are couplable to the common control module 610. For example, the memory 624 may include program instructions for controlling the firing motor 602, the closure motor 603, and the articulation motors 606 a, 606 b. Such program instructions may cause the processor 622 to control the firing, closure, and articulation functions in accordance with inputs from algorithms or control programs of the surgical instrument or tool.

One or more mechanisms and/or sensors such as, for example, sensors 630 can be employed to alert the processor 622 to the program instructions that should be used in a particular setting. For example, the sensors 630 may alert the processor 622 to use the program instructions associated with firing, closing, and articulating the end effector. In certain instances, the sensors 630 may comprise position sensors which can be employed to sense the position of the switch 614, for example. Accordingly, the processor 622 may use the program instructions associated with firing the I-beam of the end effector upon detecting, through the sensors 630 for example, that the switch 614 is in the first position 616; the processor 622 may use the program instructions associated with closing the anvil upon detecting, through the sensors 630 for example, that the switch 614 is in the second position 617; and the processor 622 may use the program instructions associated with articulating the end effector upon detecting, through the sensors 630 for example, that the switch 614 is in the third or fourth position 618 a, 618 b.

FIG. 9 illustrates a diagram of a situationally aware surgical system 5100, in accordance with at least one aspect of the present disclosure. In some exemplifications, the data sources 5126 may include, for example, the modular devices 5102 (which can include sensors configured to detect parameters associated with the patient and/or the modular device itself), databases 5122 (e.g., an EMR database containing patient records), and patient monitoring devices 5124 (e.g., a blood pressure (BP) monitor and an electrocardiography (EKG) monitor). The surgical hub 5104 can be configured to derive the contextual information pertaining to the surgical procedure from the data based upon, for example, the particular combination(s) of received data or the particular order in which the data is received from the data sources 5126. The contextual information inferred from the received data can include, for example, the type of surgical procedure being performed, the particular step of the surgical procedure that the surgeon is performing, the type of tissue being operated on, or the body cavity that is the subject of the procedure. This ability by some aspects of the surgical hub 5104 to derive or infer information related to the surgical procedure from received data can be referred to as “situational awareness.” In an exemplification, the surgical hub 5104 can incorporate a situational awareness system, which is the hardware and/or programming associated with the surgical hub 5104 that derives contextual information pertaining to the surgical procedure from the received data.

The situational awareness system of the surgical hub 5104 can be configured to derive the contextual information from the data received from the data sources 5126 in a variety of different ways. In an exemplification, the situational awareness system can include a pattern recognition system, or machine learning system (e.g., an artificial neural network), that has been trained on training data to correlate various inputs (e.g., data from databases 5122, patient monitoring devices 5124, and/or modular devices 5102) to corresponding contextual information regarding a surgical procedure. In other words, a machine learning system can be trained to accurately derive contextual information regarding a surgical procedure from the provided inputs. In examples, the situational awareness system can include a lookup table storing pre-characterized contextual information regarding a surgical procedure in association with one or more inputs (or ranges of inputs) corresponding to the contextual information. In response to a query with one or more inputs, the lookup table can return the corresponding contextual information for the situational awareness system for controlling the modular devices 5102. In examples, the contextual information received by the situational awareness system of the surgical hub 5104 can be associated with a particular control adjustment or set of control adjustments for one or more modular devices 5102. In examples, the situational awareness system can include a further machine learning system, lookup table, or other such system, which generates or retrieves one or more control adjustments for one or more modular devices 5102 when provided the contextual information as input.

A surgical hub 5104 incorporating a situational awareness system can provide a number of benefits for the surgical system 5100. One benefit may include improving the interpretation of sensed and collected data, which would in turn improve the processing accuracy and/or the usage of the data during the course of a surgical procedure. To return to a previous example, a situationally aware surgical hub 5104 could determine what type of tissue was being operated on; therefore, when an unexpectedly high force to close the surgical instrument's end effector is detected, the situationally aware surgical hub 5104 could correctly ramp up or ramp down the motor of the surgical instrument for the type of tissue.

The type of tissue being operated can affect the adjustments that are made to the compression rate and load thresholds of a surgical stapling and cutting instrument for a particular tissue gap measurement. A situationally aware surgical hub 5104 could infer whether a surgical procedure being performed is a thoracic or an abdominal procedure, allowing the surgical hub 5104 to determine whether the tissue clamped by an end effector of the surgical stapling and cutting instrument is lung (for a thoracic procedure) or stomach (for an abdominal procedure) tissue. The surgical hub 5104 could then adjust the compression rate and load thresholds of the surgical stapling and cutting instrument appropriately for the type of tissue.

The type of body cavity being operated in during an insufflation procedure can affect the function of a smoke evacuator. A situationally aware surgical huh 5104 could determine whether the surgical site is under pressure (by determining that the surgical procedure is utilizing insufflation) and determine the procedure type. As a procedure type can be generally performed in a specific body cavity, the surgical hub 5104 could then control the motor rate of the smoke evacuator appropriately for the body cavity being operated in. Thus, a situationally aware surgical hub 5104 could provide a consistent amount of smoke evacuation for both thoracic and abdominal procedures.

The type of procedure being performed can affect the optimal energy level for an ultrasonic surgical instrument or radio frequency (RF) electrosurgical instrument to operate at. Arthroscopic procedures, for example, may require higher energy levels because the end effector of the ultrasonic surgical instrument or RF electrosurgical instrument is immersed in fluid. A situationally aware surgical hub 5104 could determine whether the surgical procedure is an arthroscopic procedure. The surgical hub 5104 could then adjust the RF power level or the ultrasonic amplitude of the generator (i.e., “energy level”) to compensate for the fluid filled environment. Relatedly, the type of tissue being operated on can affect the optimal energy level for an ultrasonic surgical instrument or RF electrosurgical instrument to operate at. A situationally aware surgical hub 5104 could determine what type of surgical procedure is being performed and then customize the energy level for the ultrasonic surgical instrument or RF electrosurgical instrument, respectively, according to the expected tissue profile for the surgical procedure. Furthermore, a situationally aware surgical hub 5104 can be configured to adjust the energy level for the ultrasonic surgical instrument or RF electrosurgical instrument throughout the course of a surgical procedure, rather than just on a procedure-by-procedure basis. A situationally aware surgical hub 5104 could determine what step of the surgical procedure is being performed or will subsequently be performed and then update the control algorithms for the generator and/or ultrasonic surgical instrument or RF electrosurgical instrument to set the energy level at a value appropriate for the expected tissue type according to the surgical procedure step.

In examples, data can be drawn from additional data sources 5126 to improve the conclusions that the surgical hub 5104 draws from one data source 5126. A situationally aware surgical hub 5104 could augment data that it receives from the modular devices 5102 with contextual information that it has built up regarding the surgical procedure from other data sources 5126. For example, a situationally aware surgical hub 5104 can be configured to determine whether hemostasis has occurred (i.e., whether bleeding at a surgical site has stopped) according to video or image data received from a medical imaging device. However, in some cases the video or image data can be inconclusive. Therefore, in an exemplification, the surgical hub 5104 can be further configured to compare a physiologic measurement (e.g., blood pressure sensed by a BP monitor communicably connected to the surgical hub 5104) with the visual or image data of hemostasis (e.g., from a medical imaging device 124 (FIG. 2 ) communicably coupled to the surgical hub 5104) to make a determination on the integrity of the staple line or tissue weld. In other words, the situational awareness system of the surgical hub 5104 can consider the physiological measurement data to provide additional context in analyzing the visualization data. The additional context can be useful when the visualization data may be inconclusive or incomplete on its own.

For example, a situationally aware surgical hub 5104 could proactively activate the generator to which an RF electrosurgical instrument is connected if it determines that a subsequent step of the procedure requires the use of the instrument. Proactively activating the energy source can allow the instrument to be ready for use a soon as the preceding step of the procedure is completed.

The situationally aware surgical hub 5104 could determine whether the current or subsequent step of the surgical procedure requires a different view or degree of magnification on the display according to the feature(s) at the surgical site that the surgeon is expected to need to view. The surgical hub 5104 could then proactively change the displayed view (supplied by, e.g., a medical imaging device for the visualization system 108) accordingly so that the display automatically adjusts throughout the surgical procedure.

The situationally aware surgical hub 5104 could determine which step of the surgical procedure is being performed or will subsequently be performed and whether particular data or comparisons between data will be required for that step of the surgical procedure. The surgical hub 5104 can be configured to automatically call up data screens based upon the step of the surgical procedure being performed, without waiting for the surgeon to ask for the particular information.

Errors may be checked during the setup of the surgical procedure or during the course of the surgical procedure. For example, the situationally aware surgical hub 5104 could determine whether the operating theater is setup properly or optimally for the surgical procedure to be performed. The surgical hub 5104 can be configured to determine the type of surgical procedure being performed, retrieve the corresponding checklists, product location, or setup needs (e.g., from a memory), and then compare the current operating theater layout to the standard layout for the type of surgical procedure that the surgical hub 5104 determines is being performed. In some exemplifications, the surgical hub 5104 can be configured to compare the list of items for the procedure and/or a list of devices paired with the surgical hub 5104 to a recommended or anticipated manifest of items and/or devices for the given surgical procedure. If there are any discontinuities between the lists, the surgical hub 5104 can be configured to provide an alert indicating that a particular modular device 5102, patient monitoring device 5124, and/or other surgical item is missing. In some exemplifications, the surgical hub 5104 can be configured to determine the relative distance or position of the modular devices 5102 and patient monitoring devices 5124 via proximity sensors, for example. The surgical hub 5104 can compare the relative positions of the devices to a recommended or anticipated layout for the particular surgical procedure. If there are any discontinuities between the layouts, the surgical hub 5104 can be configured to provide an alert indicating that the current layout for the surgical procedure deviates from the recommended layout.

The situationally aware surgical hub 5104 could determine whether the surgeon (or other medical personnel) was making an error or otherwise deviating from the expected course of action during the course of a surgical procedure. For example, the surgical hub 5104 can be configured to determine the type of surgical procedure being performed, retrieve the corresponding list of steps or order of equipment usage (e.g., from a memory), and then compare the steps being performed or the equipment being used during the course of the surgical procedure to the expected steps or equipment for the type of surgical procedure that the surgical hub 5104 determined is being performed. In some exemplifications, the surgical hub 5104 can be configured to provide an alert indicating that an unexpected action is being performed or an unexpected device is being utilized at the particular step in the surgical procedure.

The surgical instruments (and other modular devices 5102) may be adjusted for the particular context of each surgical procedure (such as adjusting to different tissue types) and validating actions during a surgical procedure. Next steps, data, and display adjustments may be provided to surgical instruments (and other modular devices 5102) in the surgical theater according to the specific context of the procedure.

FIG. 10 illustrates a timeline 5200 of an illustrative surgical procedure and the contextual information that a surgical hub 5104 can derive from the data received from the data sources 5126 at each step in the surgical procedure. In the following description of the timeline 5200 illustrated in FIG. 9 , reference should also be made to FIG. 9 . The timeline 5200 may depict the typical steps that would be taken by the nurses, surgeons, and other medical personnel during the course of a lung segmentectomy procedure, beginning with setting up the operating theater and ending with transferring the patient to a post-operative recovery room. The situationally aware surgical hub 5104 may receive data from the data sources 5126 throughout the course of the surgical procedure, including data generated each time medical personnel utilize a modular device 5102 that is paired with the surgical hub 5104. The surgical hub 5104 can receive this data from the paired modular devices 5102 and other data sources 5126 and continually derive inferences (i.e., contextual information) about the ongoing procedure as new data is received, such as which step of the procedure is being performed at any given time. The situational awareness system of the surgical hub 5104 can be able to, for example, record data pertaining to the procedure for generating reports, verify the steps being taken by the medical personnel, provide data or prompts (e.g., via a display screen) that may be pertinent for the particular procedural step, adjust modular devices 5102 based on the context (e.g., activate monitors, adjust the FOV of the medical imaging device, or change the energy level of an ultrasonic surgical instrument or RF electrosurgical instrument), and take any other such action described herein.

As the first step 5202 in this illustrative procedure, the hospital staff members may retrieve the patient's EMR from the hospital's EMR database. Based on select patient data in the EMR, the surgical hub 5104 determines that the procedure to be performed is a thoracic procedure. Second 5204, the staff members may scan the incoming medical supplies for the procedure. The surgical hub 5104 cross-references the scanned supplies with a list of supplies that can be utilized in various types of procedures and confirms that the mix of supplies corresponds to a thoracic procedure. Further, the surgical hub 5104 may also be able to determine that the procedure is not a wedge procedure (because the incoming supplies either lack certain supplies that are necessary for a thoracic wedge procedure or do not otherwise correspond to a thoracic wedge procedure). Third 5206, the medical personnel may scan the patient band via a scanner 5128 that is communicably connected to the surgical hub 5104. The surgical hub 5104 can then confirm the patient's identity based on the scanned data. Fourth 5208, the medical staff turns on the auxiliary equipment. The auxiliary equipment being utilized can vary according to the type of surgical procedure and the techniques to be used by the surgeon, but in this illustrative case they include a smoke evacuator, insufflator, and medical imaging device. When activated, the auxiliary equipment that are modular devices 5102 can automatically pair with the surgical hub 5104 that may be located within a particular vicinity of the modular devices 5102 as part of their initialization process. The surgical hub 5104 can then derive contextual information about the surgical procedure by detecting the types of modular devices 5102 that pair with it during this pre-operative or initialization phase. In this particular example, the surgical hub 5104 may determine that the surgical procedure is a VATS procedure based on this particular combination of paired modular devices 5102. Based on the combination of the data from the patient's EMR, the list of medical supplies to be used in the procedure, and the type of modular devices 5102 that connect to the hub, the surgical hub 5104 can generally infer the specific procedure that the surgical team will be performing. Once the surgical hub 5104 knows what specific procedure is being performed, the surgical hub 5104 can then retrieve the steps of that procedure from a memory or from the cloud and then cross-reference the data it subsequently receives from the connected data sources 5126 (e.g., modular devices 5102 and patient monitoring devices 5124) to infer what step of the surgical procedure the surgical team is performing. Fifth 5210, the staff members attach the FKG electrodes and other patient monitoring devices 5124 to the patient. The KKG electrodes and other patient monitoring devices 5124 may pair with the surgical hub 5104. As the surgical hub 5104 begins receiving data from the patient monitoring devices 5124, the surgical hub 5104 may confirm that the patient is in the operating theater, as described in the process 5207, for example. Sixth 5212, the medical personnel may induce anesthesia in the patient. The surgical hub 5104 can infer that the patient is under anesthesia based on data from the modular devices 5102 and/or patient monitoring devices 5124, including EKG data, blood pressure data, ventilator data, or combinations thereof, for example. Upon completion of the sixth step 5212, the pre-operative portion of the lung segmentectomy procedure is completed and the operative portion begins.

Seventh 5214, the patient's lung that is being operated on may be collapsed (while ventilation is switched to the contralateral lung). The surgical hub 5104 can infer from the ventilator data that the patient's lung has been collapsed, for example. The surgical hub 5104 can infer that the operative portion of the procedure has commenced as it can compare the detection of the patient's lung collapsing to the expected steps of the procedure (which can be accessed or retrieved previously) and thereby determine that collapsing the lung can be the first operative step in this particular procedure. Eighth 5216, the medical imaging device 5108 (e.g., a scope) may be inserted and video from the medical imaging device may be initiated. The surgical hub 5104 may receive the medical imaging device data (i.e., video or image data) through its connection to the medical imaging device. Upon receipt of the medical imaging device data, the surgical hub 5104 can determine that the laparoscopic portion of the surgical procedure has commenced. Further, the surgical hub 5104 can determine that the particular procedure being performed is a segmentectomy, as opposed to a lobectomy (note that a wedge procedure has already been discounted by the surgical hub 5104 based on data received at the second step 5204 of the procedure). The data from the medical imaging device 124 (FIG. 2 ) can be utilized to determine contextual information regarding the type of procedure being performed in a number of different ways, including by determining the angle at which the medical imaging device is oriented with respect to the visualization of the patient's anatomy, monitoring the number or medical imaging devices being utilized (i.e., that are activated and paired with the surgical hub 5104), and monitoring the types of visualization devices utilized. For example, one technique for performing a VATS lobectomy may place the camera in the lower anterior corner of the patient's chest cavity above the diaphragm, whereas one technique for performing a VATS segmentectomy places the camera in an anterior intercostal position relative to the segmental fissure. Using pattern recognition or machine learning techniques, for example, the situational awareness system can be trained to recognize the positioning of the medical imaging device according to the visualization of the patient's anatomy. An example technique for performing a VATS lobectomy may utilize a single medical imaging device. An example technique for performing a VATS segmentectomy utilizes multiple cameras. An example technique for performing a VATS segmentectomy utilizes an infrared light source (which can be communicably coupled to the surgical hub as part of the visualization system) to visualize the segmental fissure, which is not utilized in a VATS lobectomy. By tracking any or all of this data from the medical imaging device 5108, the surgical hub 5104 can thereby determine the specific type of surgical procedure being performed and/or the technique being used for a particular type of surgical procedure.

Ninth 5218, the surgical team may begin the dissection step of the procedure. The surgical hub 5104 can infer that the surgeon is in the process of dissecting to mobilize the patient's lung because it receives data from the RF or ultrasonic generator indicating that an energy instrument is being fired. The surgical hub 5104 can cross-reference the received data with the retrieved steps of the surgical procedure to determine that an energy instrument being fired at this point in the process (i.e., after the completion of the previously discussed steps of the procedure) corresponds to the dissection step. Tenth 5220, the surgical team may proceed to the ligation step of the procedure. The surgical hub 5104 can infer that the surgeon is ligating arteries and veins because it may receive data from the surgical stapling and cutting instrument indicating that the instrument is being fired. Similar to the prior step, the surgical hub 5104 can derive this inference by cross-referencing the receipt of data from the surgical stapling and cutting instrument with the retrieved steps in the process. Eleventh 5222, the segmentectomy portion of the procedure can be performed. The surgical hub 5104 can infer that the surgeon is transecting the parenchyma based on data from the surgical stapling and cutting instrument, including data from its cartridge. The cartridge data can correspond to the size or type of staple being fired by the instrument, for example. As different types of staples are utilized for different types of tissues, the cartridge data can thus indicate the type of tissue being stapled and/or transected. In this case, the type of staple being fired is utilized for parenchyma (or other similar tissue types), which allows the surgical hub 5104 to infer that the segmentectomy portion of the procedure is being performed. Twelfth 5224, the node dissection step is then performed. The surgical hub 5104 can infer that the surgical team is dissecting the node and performing a leak test based on data received from the generator indicating that an RF or ultrasonic instrument is being fired. For this particular procedure, an RF or ultrasonic instrument being utilized after parenchyma was transected corresponds to the node dissection step, which allows the surgical hub 5104 to make this inference. It should be noted that surgeons regularly switch hack and forth between surgical stapling/cutting instruments and surgical energy (e.g., RF or ultrasonic) instruments depending upon the particular step in the procedure because different instruments are better adapted for particular tasks. Therefore, the particular sequence in which the stapling/cutting instruments and surgical energy instruments are used can indicate what step of the procedure the surgeon is performing. Upon completion of the twelfth step 5224, the incisions and closed up and the post-operative portion of the procedure may begin.

Thirteenth 5226, the patient's anesthesia can be reversed. The surgical hub 5104 can infer that the patient is emerging from the anesthesia based on the ventilator data (i.e., the patient's breathing rate begins increasing), for example. Lastly, the fourteenth step 5228 may be that the medical personnel remove the various patient monitoring devices 5124 from the patient. The surgical hub 5104 can thus infer that the patient is being transferred to a recovery room when the hub loses EKG, BP, and other data from the patient monitoring devices 5124. As can be seen from the description of this illustrative procedure, the surgical hub 5104 can determine or infer when each step of a given surgical procedure is taking place according to data received from the various data sources 5126 that are communicably coupled to the surgical hub 5104.

In addition to utilizing the patient data from EMR database(s) to infer the type of surgical procedure that is to be performed, as illustrated in the first step 5202 of the timeline 5200 depicted in FIG. 10 , the patient data can also be utilized by a situation-ally aware surgical hub 5104 to generate control adjustments for the paired modular devices 5102.

FIG. 11 is a block diagram of the computer-implemented interactive surgical system, in accordance with at least one aspect of the present disclosure. In one aspect, the computer-implemented interactive surgical system may be configured to monitor and analyze data related to the operation of various surgical systems that include surgical hubs, surgical instruments, robotic devices and operating theaters or healthcare facilities. The computer-implemented interactive surgical system may comprise a cloud-based analytics system. Although the cloud-based analytics system may be described as a surgical system, it may not be necessarily limited as such and could be a cloud-based medical system generally. As illustrated in FIG. 11 , the cloud-based analytics system may comprise a plurality of surgical instruments 7012 (may be the same or similar to instruments 112), a plurality of surgical hubs 7006 (may be the same or similar to hubs 106), and a surgical data network 7001 (may be the same or similar to network 201) to couple the surgical hubs 7006 to the cloud 7004 (may be the same or similar to cloud 204). Each of the plurality of surgical hubs 7006 may be communicatively coupled to one or more surgical instruments 7012. The hubs 7006 may also be communicatively coupled to the cloud 7004 of the computer-implemented interactive surgical system via the network 7001. The cloud 7004 may be a remote centralized source of hardware and software for storing, manipulating, and communicating data generated based on the operation of various surgical systems. As shown in FIG. 11 , access to the cloud 7004 may be achieved via the network 7001, which may be the Internet or some other suitable computer network. Surgical hubs 7006 that may be coupled to the cloud 7004 can be considered the client side of the cloud computing system (i.e., cloud-based analytics system). Surgical instruments 7012 may be paired with the surgical hubs 7006 for control and implementation of various surgical procedures or operations as described herein.

In addition, surgical instruments 7012 may comprise transceivers for data transmission to and from their corresponding surgical hubs 7006 (which may also comprise transceivers). Combinations of surgical instruments 7012 and corresponding hubs 7006 may indicate particular locations, such as operating theaters in healthcare facilities (e.g., hospitals), for providing medical operations. For example, the memory of a surgical hub 7006 may store location data. As shown in FIG. 11 , the cloud 7004 comprises central servers 7013 (may be same or similar to remote server 7013), hub application servers 7002, data analytics modules 7034, and an input/output (“I/O”) interface 7006. The central servers 7013 of the cloud 7004 collectively administer the cloud computing system, which includes monitoring requests by client surgical hubs 7006 and managing the processing capacity of the cloud 7004 for executing the requests. Each of the central servers 7013 may comprise one or more processors 7008 coupled to suitable memory devices 7010 which can include volatile memory such as random-access memory (RAM) and non-volatile memory such as magnetic storage devices. The memory devices 7010 may comprise machine executable instructions that when executed cause the processors 7008 to execute the data analytics modules 7034 for the cloud-based data analysis, operations, recommendations and other operations described below. Moreover, the processors 7008 can execute the data analytics modules 7034 independently or in conjunction with hub applications independently executed by the hubs 7006. The central servers 7013 also may comprise aggregated medical data databases 2212, which can reside in the memory 2210.

Based on connections to various surgical hubs 7006 via the network 7001, the cloud 7004 can aggregate data from specific data generated by various surgical instruments 7012 and their corresponding hubs 7006. Such aggregated data may be stored within the aggregated medical databases 7012 of the cloud 7004. In particular, the cloud 7004 may advantageously perform data analysis and operations on the aggregated data to yield insights and/or perform functions that individual hubs 7006 could not achieve on their own. To this end, as shown in FIG. 11 , the cloud 7004 and the surgical hubs 7006 are communicatively coupled to transmit and receive information. The I/O interface 7006 is connected to the plurality of surgical hubs 7006 via the network 7001. In this way, the I/O interface 7006 can be configured to transfer information between the surgical hubs 7006 and the aggregated medical data databases 7011. Accordingly, the I/O interface 7006 may facilitate read/write operations of the cloud-based analytics system. Such read/write operations may be executed in response to requests from hubs 7006. These requests could be transmitted to the hubs 7006 through the hub applications. The I/O interface 7006 may include one or more high speed data ports, which may include universal serial bus (USB) ports, IEEE 1394 ports, as well as Wi-Fi and Bluetooth I/O interfaces for connecting the cloud 7004 to hubs 7006. The hub application servers 7002 of the cloud 7004 may be configured to host and supply shared capabilities to software applications (e.g., hub applications) executed by surgical hubs 7006. For example, the huh application servers 7002 may manage requests made by the hub applications through the hubs 7006, control access to the aggregated medical data databases 7011, and perform load balancing. The data analytics modules 7034 are described in further detail with reference to FIG. 12 .

The particular cloud computing system configuration described in the present disclosure may be specifically designed to address various issues arising in the context of medical operations and procedures performed using medical devices, such as the surgical instruments 7012, 112. In particular, the surgical instruments 7012 may be digital surgical devices configured to interact with the cloud 7004 for implementing techniques to improve the performance of surgical operations. Various surgical instruments 7012 and/or surgical hubs 7006 may comprise touch-controlled user interfaces such that clinicians may control aspects of interaction between the surgical instruments 7012 and the cloud 7004. Other suitable user interfaces for control such as auditory controlled user interfaces can also be used.

FIG. 12 is a block diagram which illustrates the functional architecture of the computer-implemented interactive surgical system, in accordance with at least one aspect of the present disclosure. The cloud-based analytics system may include a plurality of data analytics modules 7034 that may be executed by the processors 7008 of the cloud 7004 for providing data analytic solutions to problems specifically arising in the medical field. As shown in FIG. 12 , the functions of the cloud-based data analytics modules 7034 may be assisted via hub applications 7014 hosted by the hub application servers 7002 that may be accessed on surgical hubs 7006. The cloud processors 7008 and huh applications 7014 may operate in conjunction to execute the data analytics modules 7034. Application program interfaces (APIs) 7016 may define the set of protocols and routines corresponding to the hub applications 7014. Additionally, the APIs 7016 may manage the storing and retrieval of data into and from the aggregated medical databases 7012 for the operations of the applications 7014. The caches 7018 may also store data (e.g., temporarily) and may be coupled to the APIs 7016 for more efficient retrieval of data used by the applications 7014. The data analytics modules 7034 in FIG. 12 may include modules for resource optimization 7020, data collection and aggregation 7022, authorization and security 7024, control program updating 7026, patient outcome analysis 7028, recommendations 7030, and data sorting and prioritization 7032. Other suitable data analytics modules could also be implemented by the cloud 7004, according to some aspects. In one aspect, the data analytics modules may be used for specific recommendations based on analyzing trends, outcomes, and other data.

For example, the data collection and aggregation module 7022 could be used to generate self-describing data (e.g., metadata) including identification of notable features or configuration (e.g., trends), management of redundant data sets, and storage of the data in paired data sets which can be grouped by surgery but not necessarily keyed to actual surgical dates and surgeons. In particular, pair data sets generated from operations of surgical instruments 7012 can comprise applying a binary classification, e.g., a bleeding or a non-bleeding event. More generally, the binary classification may be characterized as either a desirable event (e.g., a successful surgical procedure) or an undesirable event (e.g., a misfired or misused surgical instrument 7012). The aggregated self-describing data may correspond to individual data received from various groups or subgroups of surgical hubs 7006. Accordingly, the data collection and aggregation module 7022 can generate aggregated metadata or other organized data based on raw data received from the surgical hubs 7006. To this end, the processors 7008 can be operationally coupled to the hub applications 7014 and aggregated medical data databases 7011 for executing the data analytics modules 7034. The data collection and aggregation module 7022 may store the aggregated organized data into the aggregated medical data databases 2212.

The resource optimization module 7020 can be configured to analyze this aggregated data to determine an optimal usage of resources for a particular or group of healthcare facilities. For example, the resource optimization module 7020 may determine an optimal order point of surgical stapling instruments 7012 for a group of healthcare facilities based on corresponding predicted demand of such instruments 7012. The resource optimization module 7020 might also assess the resource usage or other operational configurations of various healthcare facilities to determine whether resource usage could be improved. Similarly, the recommendations module 7030 can be configured to analyze aggregated organized data from the data collection and aggregation module 7022 to provide recommendations. For example, the recommendations module 7030 could recommend to healthcare facilities (e.g., medical service providers such as hospitals) that a particular surgical instrument 7012 should be upgraded to an improved version based on a higher than expected error rate, for example. Additionally, the recommendations module 7030 and/or resource optimization module 7020 could recommend better supply chain parameters such as product reorder points and provide suggestions of different surgical instrument 7012, uses thereof, or procedure steps to improve surgical outcomes. The healthcare facilities can receive such recommendations via corresponding surgical hubs 7006. More specific recommendations regarding parameters or configurations of various surgical instruments 7012 can also be provided. Hubs 7006 and/or surgical instruments 7012 each could also have display screens that display data or recommendations provided by the cloud 7004.

The patient outcome analysis module 7028 can analyze surgical outcomes associated with currently used operational parameters of surgical instruments 7012. The patient outcome analysis module 7028 may also analyze and assess other potential operational parameters. In this connection, the recommendations module 7030 could recommend using these other potential operational parameters based on yielding better surgical outcomes, such as better sealing or less bleeding. For example, the recommendations module 7030 could transmit recommendations to a surgical 7006 regarding when to use a particular cartridge for a corresponding stapling surgical instrument 7012. Thus, the cloud-based analytics system, while controlling for common variables, may be configured to analyze the large collection of raw data and to provide centralized recommendations over multiple healthcare facilities (advantageously determined based on aggregated data). For example, the cloud-based analytics system could analyze, evaluate, and/or aggregate data based on type of medical practice, type of patient, number of patients, geographic similarity between medical providers, which medical providers/facilities use similar types of instruments, etc., in a way that no single healthcare facility alone would be able to analyze independently. The control program updating module 7026 could be configured to implement various surgical instrument 7012 recommendations when corresponding control programs are updated. For example, the patient outcome analysis module 7028 could identify correlations linking specific control parameters with successful (or unsuccessful) results.

Such correlations may be addressed when updated control programs are transmitted to surgical instruments 7012 via the control program updating module 7026. Updates to instruments 7012 that may be transmitted via a corresponding hub 7006 may incorporate aggregated performance data that was gathered and analyzed by the data collection and aggregation module 7022 of the cloud 7004. Additionally, the patient outcome analysis module 7028 and recommendations module 7030 could identify improved methods of using instruments 7012 based on aggregated performance data.

The cloud-based analytics system may include security features implemented by the cloud 7004. These security features may be managed by the authorization and security module 7024. Each surgical hub 7006 can have associated unique credentials such as username, password, and other suitable security credentials. These credentials could be stored in the memory 7010 and be associated with a permitted cloud access level. For example, based on providing accurate credentials, a surgical hub 7006 may be granted access to communicate with the cloud to a predetermined extent (e.g., may only engage in transmitting or receiving certain defined types of information). To this end, the aggregated medical data databases 7011 of the cloud 7004 may comprise a database of authorized credentials for verifying the accuracy of provided credentials. Different credentials may be associated with varying levels of permission for interaction with the cloud 7004, such as a predetermined access level for receiving the data analytics generated by the cloud 7004. Furthermore, for security purposes, the cloud could maintain a database of hubs 7006, instruments 7012, and other devices that may comprise a “black list” of prohibited devices. In particular, a surgical hubs 7006 listed on the black list may not be permitted to interact with the cloud, while surgical instruments 7012 listed on the black list may not have functional access to a corresponding hub 7006 and/or may be prevented from fully functioning when paired to its corresponding hub 7006. Additionally, or alternatively, the cloud 7004 may flag instruments 7012 based on incompatibility or other specified criteria. In this manner, counterfeit medical devices and improper reuse of such devices throughout the cloud-based analytics system can be identified and addressed.

The surgical instruments 7012 may use wireless transceivers to transmit wireless signals that may represent, for example, authorization credentials for access to corresponding hubs 7006 and the cloud 7004. Wired transceivers may also be used to transmit signals. Such authorization credentials can be stored in the respective memory devices of the surgical instruments 7012. The authorization and security module 7024 can determine whether the authorization credentials are accurate or counterfeit. The authorization and security module 7024 may also dynamically generate authorization credentials for enhanced security. The credentials could also be encrypted, such as by using hash-based encryption. Upon transmitting proper authorization, the surgical instruments 7012 may transmit a signal to the corresponding hubs 7006 and ultimately the cloud 7004 to indicate that the instruments 7012 are ready to obtain and transmit medical data. In response, the cloud 7004 may transition into a state enabled for receiving medical data for storage into the aggregated medical data databases 7011. This data transmission readiness could be indicated by a light indicator on the instruments 7012, for example. The cloud 7004 can also transmit signals to surgical instruments 7012 for updating their associated control programs. The cloud 7004 can transmit signals that are directed to a particular class of surgical instruments 7012 (e.g., electrosurgical instruments) so that software updates to control programs are only transmitted to the appropriate surgical instruments 7012. Moreover, the cloud 7004 could be used to implement system wide solutions to address local or global problems based on selective data transmission and authorization credentials. For example, if a group of surgical instruments 7012 are identified as having a common manufacturing defect, the cloud 7004 may change the authorization credentials corresponding to this group to implement an operational lockout of the group.

The cloud-based analytics system may allow for monitoring multiple healthcare facilities (e.g., medical facilities like hospitals) to determine improved practices and recommend changes (via the recommendations module 2030, for example) accordingly. Thus, the processors 7008 of the cloud 7004 can analyze data associated with an individual healthcare facility to identify the facility and aggregate the data with other data associated with other healthcare facilities in a group. Groups could be defined based on similar operating practices or geographical location, for example. In this way, the cloud 7004 may provide healthcare facility group wide analysis and recommendations. The cloud-based analytics system could also be used for enhanced situational awareness. For example, the processors 7008 may predictively model the effects of recommendations on the cost and effectiveness for a particular facility (relative to overall operations and/or various medical procedures). The cost and effectiveness associated with that particular facility can also be compared to a corresponding local region of other facilities or any other comparable facilities.

The data sorting and prioritization module 7032 may prioritize and sort data based on criticality (e.g., the severity of a medical event associated with the data, unexpectedness, suspiciousness). This sorting and prioritization may be used in conjunction with the functions of the other data analytics modules 7034 described herein to improve the cloud-based analytics and operations described herein. For example, the data sorting and prioritization module 7032 can assign a priority to the data analysis performed by the data collection and aggregation module 7022 and patient outcome analysis modules 7028. Different prioritization levels can result in particular responses from the cloud 7004 (corresponding to a level of urgency) such as escalation for an expedited response, special processing, exclusion from the aggregated medical data databases 7011, or other suitable responses. Moreover, if necessary, the cloud 7004 can transmit a request (e.g., a push message) through the hub application servers for additional data from corresponding surgical instruments 7012. The push message can result in a notification displayed on the corresponding hubs 7006 for requesting supporting or additional data. This push message may be required in situations in which the cloud detects a significant irregularity or outlier and the cloud cannot determine the cause of the irregularity. The central servers 7013 may be programmed to trigger this push message in certain significant circumstances, such as when data is determined to be different from an expected value beyond a predetermined threshold or when it appears security has been comprised, for example.

Additional example details for the various functions described are provided in the ensuing descriptions below. Each of the various descriptions may utilize the cloud architecture as described in FIGS. 11 and 12 as one example of hardware and software implementation.

FIG. 13 illustrates a block diagram of a computer-implemented adaptive surgical system 9060 that is configured to adaptively generate control program updates for modular devices 9050, in accordance with at least one aspect of the present disclosure. In some exemplifications, the surgical system may include a surgical hub 9000, multiple modular devices 9050 communicably coupled to the surgical hub 9000, and an analytics system 9100 communicably coupled to the surgical hub 9000. Although a single surgical hub 9000 may be depicted, it should be noted that the surgical system 9060 can include any number of surgical hubs 9000, which can be connected to form a network of surgical hubs 9000 that are communicably coupled to the analytics system 9010. In some exemplifications, the surgical hub 9000 may include a processor 9010 coupled to a memory 9020 for executing instructions stored thereon and a data relay interface 9030 through which data is transmitted to the analytics system 9100. In some exemplifications, the surgical hub 9000 further may include a user interface 9090 having an input device 9092 (e.g., a capacitive touchscreen or a keyboard) for receiving inputs from a user and an output device 9094 (e.g., a display screen) for providing outputs to a user. Outputs can include data from a query input by the user, suggestions for products or mixes of products to use in a given procedure, and/or instructions for actions to be carried out before, during, or after surgical procedures. The surgical hub 9000 further may include an interface 9040 for communicably coupling the modular devices 9050 to the surgical hub 9000. In one aspect, the interface 9040 may include a transceiver that is communicably connectable to the modular device 9050 via a wireless communication protocol. The modular devices 9050 can include, for example, surgical stapling and cutting instruments, electrosurgical instruments, ultrasonic instruments, insufflators, respirators, and display screens. In some exemplifications, the surgical hub 9000 can further be communicably coupled to one or more patient monitoring devices 9052, such as EKG monitors or BP monitors. In some exemplifications, the surgical hub 9000 can further be communicably coupled to one or more databases 9054 or external computer systems, such as an EMR database of the medical facility at which the surgical hub 9000 is located.

When the modular devices 9050 are connected to the surgical hub 9000, the surgical hub 9000 can sense or receive perioperative data from the modular devices 9050 and then associate the received perioperative data with surgical procedural outcome data. The perioperative data may indicate how the modular devices 9050 were controlled during the course of a surgical procedure. The procedural outcome data includes data associated with a result from the surgical procedure (or a step thereof), which can include whether the surgical procedure (or a step thereof) had a positive or negative outcome. For example, the outcome data could include whether a patient suffered from postoperative complications from a particular procedure or whether there was leakage (e.g., bleeding or air leakage) at a particular staple or incision line. The surgical hub 9000 can obtain the surgical procedural outcome data by receiving the data from an external source (e.g., from an EMR database 9054), by directly detecting the outcome (e.g., via one of the connected modular devices 9050), or inferring the occurrence of the outcomes through a situational awareness system. For example, data regarding postoperative complications could be retrieved from an EMR database 9054 and data regarding staple or incision line leakages could be directly detected or inferred by a situational awareness system. The surgical procedural outcome data can be inferred by a situational awareness system from data received from a variety of data sources, including the modular devices 9050 themselves, the patient monitoring device 9052, and the databases 9054 to which the surgical huh 9000 is connected.

The surgical hub 9000 can transmit the associated modular device 9050 data and outcome data to the analytics system 9100 for processing thereon. By transmitting both the perioperative data indicating how the modular devices 9050 are controlled and the procedural outcome data, the analytics system 9100 can correlate the different manners of controlling the modular devices 9050 with surgical outcomes for the particular procedure type. In some exemplifications, the analytics system 9100 may include a network of analytics servers 9070 that are configured to receive data from the surgical hubs 9000. Each of the analytics servers 9070 can include a memory and a processor coupled to the memory that is executing instructions stored thereon to analyze the received data. In some exemplifications, the analytics servers 9070 may be connected in a distributed computing architecture and/or utilize a cloud computing architecture. Based on this paired data, the analytics system 9100 can then learn optimal or preferred operating parameters for the various types of modular devices 9050, generate adjustments to the control programs of the modular devices 9050 in the field, and then transmit (or “push”) updates to the modular devices' 9050 control programs.

Additional detail regarding the computer-implemented interactive surgical system 9060, including the surgical hub 9000 and various modular devices 9050 connectable thereto, are described in connection with FIGS. 5-6 .

FIG. 14 provides a surgical system 6500 in accordance with the present disclosure and may include a surgical instrument 6502 that can be in communication with a console 6522 or a portable device 6526 through a local area network 6518 or a cloud network 6520 via a wired or wireless connection. In various aspects, the console 6522 and the portable device 6526 may be any suitable computing device. The surgical instrument 6502 may include a handle 6504, an adapter 6508, and a loading unit 6514. The adapter 6508 releasably couples to the handle 6504 and the loading unit 6514 releasably couples to the adapter 6508 such that the adapter 6508 transmits a force from a drive shaft to the loading unit 6514. The adapter 6508 or the loading unit 6514 may include a force gauge (not explicitly shown) disposed therein to measure a force exerted on the loading unit 6514. The loading unit 6514 may include an end effector 6530 having a first jaw 6532 and a second jaw 6534. The loading unit 6514 may be an in-situ loaded or multi-firing loading unit (MFLU) that allows a clinician to fire a plurality of fasteners multiple times without requiring the loading unit 6514 to be removed from a surgical site to reload the loading unit 6514.

The first and second jaws 6532, 6534 may be configured to clamp tissue therebetween, fire fasteners through the clamped tissue, and sever the clamped tissue. The first jaw 6532 may be configured to fire at least one fastener a plurality of times, or may be configured to include a replaceable multi-fire fastener cartridge including a plurality of fasteners (e.g., staples, clips, etc.) that may be fired more than one time prior to being replaced. The second jaw 6534 may include an anvil that deforms or otherwise secures the fasteners about tissue as the fasteners are ejected from the multi-fire fastener cartridge.

The handle 6504 may include a motor that is coupled to the drive shaft to affect rotation of the drive shaft. The handle 6504 may include a control interface to selectively activate the motor. The control interface may include buttons, switches, levers, sliders, touchscreen, and any other suitable input mechanisms or user interfaces, which can be engaged by a clinician to activate the motor.

The control interface of the handle 6504 may be in communication with a controller 6528 of the handle 6504 to selectively activate the motor to affect rotation of the drive shafts. The controller 6528 may be disposed within the handle 6504 and is configured to receive input from the control interface and adapter data from the adapter 6508 or loading unit data from the loading unit 6514. The controller 6528 may analyze the input from the control interface and the data received from the adapter 6508 and/or loading unit 6514 to selectively activate the motor. The handle 6504 may also include a display that is viewable by a clinician during use of the handle 6504. The display may be configured to display portions of the adapter or loading unit data before, during, or after firing of the instrument 6502.

The adapter 6508 may include an adapter identification device 6510 disposed therein and the loading unit 6514 includes a loading unit identification device 6516 disposed therein. The adapter identification device 6510 may be in communication with the controller 6528, and the loading unit identification device 6516 may be in communication with the controller 6528. It will be appreciated that the loading unit identification device 6516 may be in communication with the adapter identification device 6510, which relays or passes communication from the loading unit identification device 6516 to the controller 6528.

The adapter 6508 may also include a plurality of sensors 6512 (one shown) disposed thereabout to detect various conditions of the adapter 6508 or of the environment (e.g., if the adapter 6508 is connected to a loading unit, if the adapter 6508 is connected to a handle, if the drive shafts are rotating, the torque of the drive shafts, the strain of the drive shafts, the temperature within the adapter 6508, a number of firings of the adapter 6508, a peak force of the adapter 6508 during firing, a total amount of force applied to the adapter 6508, a peak retraction force of the adapter 6508, a number of pauses of the adapter 6508 during firing, etc.). The plurality of sensors 6512 may provide an input to the adapter identification device 6510 in the form of data signals. The data signals of the plurality of sensors 6512 may be stored within, or be used to update the adapter data stored within, the adapter identification device 6510. The data signals of the plurality of sensors 6512 may be analog or digital. The plurality of sensors 6512 may include a force gauge to measure a force exerted on the loading unit 6514 during firing.

The handle 6504 and the adapter 6508 can be configured to interconnect the adapter identification device 6510 and the loading unit identification device 6516 with the controller 6528 via an electrical interface. The electrical interface may be a direct electrical interface (i.e., include electrical contacts that engage one another to transmit energy and signals therebetween). Additionally or alternatively, the electrical interface may be a non-contact electrical interface to wirelessly transmit energy and signals therebetween (e.g., inductively transfer). It is also contemplated that the adapter identification device 6510 and the controller 6528 may be in wireless communication with one another via a wireless connection separate from the electrical interface.

The handle 6504 may include a transmitter 6506 that is configured to transmit instrument data from the controller 6528 to other components of the system 6500 (e.g., the LAN 6518, the cloud 6520, the console 6522, or the portable device 6526). The transmitter 6506 also may receive data (e.g., cartridge data, loading unit data, or adapter data) from the other components of the system 6500. For example, the controller 6528 may transmit instrument data including a serial number of an attached adapter (e.g., adapter 6508) attached to the handle 6504, a serial number of a loading unit (e.g., loading unit 6514) attached to the adapter, and a serial number of a multi-Fire fastener cartridge (e.g., multi-fire fastener cartridge), loaded into the loading unit, to the console 6528. Thereafter, the console 6522 may transmit data (e.g., cartridge data, loading unit data, or adapter data) associated with the attached cartridge, loading unit, and adapter, respectively, back to the controller 6528. The controller 6528 can display messages on the local instrument display or transmit the message, via transmitter 6506, to the console 6522 or the portable device 6526 to display the message on the display 6524 or portable device screen, respectively.

FIG. 15A illustrates an example flow for determining a mode of operation and operating in the determined mode. The computer-implemented interactive surgical system and/or components and/or subsystems of the computer-implemented interactive surgical system may be configured to be updated. Such updates may include the inclusions of features and benefits that were not available to the user before the update. These updates may be established by any method of hardware, firmware, and software updates suitable for introducing the feature to the user. For example, replaceable/swappable (e.g., hot swappable) hardware components, flashable firmware devices, and updatable software systems may be used to update computer-implemented interactive surgical system and/or components and/or subsystems of the computer-implemented interactive surgical system.

The updates may be conditioned on any suitable criterion or set of criteria. For example, an update may be conditioned on one or more hardware capabilities of the system, such as processing capability, bandwidth, resolution, and the like. For example, the update may be conditioned on one or more software aspects, such as a purchase of certain software code. For example, the update may be conditioned on a purchased service tier. The service tier may represent a feature and/or a set of features the user is entitled to use in connection with the computer-implemented interactive surgical system. The service tier may be determined by a license code, an e-commerce server authentication interaction, a hardware key, a username/password combination, a biometric authentication interaction, a public/private key exchange interaction, or the like.

At 10704, a system/device parameter may be identified. The system/device parameter may be any element or set of elements on which an update in conditioned. For example, the computer-implemented interactive surgical system may detect a certain bandwidth of communication between a modular device and a surgical hub. For example, the computer-implemented interactive surgical system may detect an indication of the purchase of certain service tier.

At 10708, a mode of operation may be determined based on the identified system/device parameter. This determination may be made by a process that maps system/device parameters to modes of operation. The process may be a manual and/or an automated process. The process may be the result of local computation and/or remote computation. For example, a client/server interaction may be used to determine the mode of operation based on the on the identified system/device parameter. For example, local software and/or locally embedded firmware may be used to determine the mode of operation based on the identified system/device parameter. For example, a hardware key, such as a secure microprocessor for example, may be used to determine the mode of operation based on the identified system/device parameter.

At 10710, operation may proceed in accordance with the determined mode of operation. For example, a system or device may proceed to operate in a default mode of operation. For example, a system or device may proceed to operate in an alternate mode of operation. The mode of operation may be directed by control hardware, firmware, and/or software already resident in the system or device. The mode of operation may be directed by control hardware, firmware, and/or software newly installed/updated.

FIG. 15B illustrates an example functional block diagram for changing a mode of operation. An upgradeable element 10714 may include an initialization component 10716. The initialization component 10716 may include any hardware, firmware, and/or software suitable determining a mode of operation. For example, the initialization component 10716 may be portion of a system or device start-up procedure. The initialization component 10716 may engage in an interaction to determine a mode of operation for the upgradeable element 10714. For example, the initialization component 10716 may interact with a user 10730, an external resource 10732, and/or a local resource 10718 for example. For example, the initialization component 10716 may receive a licensing key from the user 10730 to determine a mode of operation. The initialization component 10716 may query an external resource 10732, such as a server for example, with a serial number of the upgradable device 10714 to determine a mode of operation. For example, the initialization component 10716 may query a local resource 10718, such as a local query to determine an amount of available bandwidth and/or a local query of a hardware key for example, to determine a mode of operation.

The upgradeable element 10714 may include one or more operation components 10720, 10722, 10726, 10728 and an operational pointer 10724. The initialization component 10716 may direct the operational pointer 10724 to direct the operation of the upgradable element 10741 to the operation component 10720, 10722, 10726, 10728 that corresponds with the determined mode of operation. The initialization component 10716 may direct the operational pointer 10724 to direct the operation of the upgradable element to a default operation component 10720. For example, the default operation component 10720 may be selected on the condition of no other alternate mode of operation being determined. For example, the default operation component 10720 may be selected on the condition of a failure of the initialization component and/or interaction failure. The initialization component 10716 may direct the operational pointer 10724 to direct the operation of the upgradable element 10714 to a resident operation component 10722. For example, certain features may be resident in the upgradable component 10714 but require activation to be put into operation. The initialization component 10716 may direct the operational pointer 10724 to direct the operation of the upgradable element 10714 to install a new operation component 10728 and/or a new installed operation component 10726. For example, new software and/or firmware may be downloaded. The new software and or firmware may contain code to enable the features represented by the selected mode of operation. For example, a new hardware component may be installed to enable the selected mode of operation.

A surgical hub may be connected, wired or wireless, with various devices and servers in the operating room, in the medical facility and/or outside of the medical facility. For example, as shown in FIG. 11 , surgical hub 7006 may be capable of communicating with surgical instruments 7012, as well as data analytic modules 7034, remote server(s) 7013 and/or hub application servers 7002. The surgical hub 7006 may communicate with the various devices and servers in different connectivity modes, which is described in more detail herein with reference to FIGS. 16A-C.

FIG. 17 shows an example flow for a hub operating under tiered communication modes. At 16104, one or more hub connectivity control parameters may be identified. At 16108, a hub connectivity mode may be determined based on the identified hub connectivity control parameter(s). For example, the surgical hub 7006 shown in FIG. 11 may determine the hub connectivity mode based on a hub connectivity control parameter. The hub connectivity mode may be selected from multiple connectivity modes that may be preconfigured, dynamically updated, semi-dynamically updated, periodically updated, or preset. The hub connectivity modes may control inter-device connectivity within a network associated with a hospital, and/or communication with an external network associated with a different hospital, for example.

The hub connectivity control parameter(s) may include, but not limited to, systems capabilities such as hardware capability, firmware capability and/or software capability. For example, if a surgical instrument lacks the hardware capability to provide indications of instructional information, the surgical hub may switch to a connectivity mode that may disable providing instructional information to the surgical instrument.

The hub connectivity control parameter(s) may include a consumer-controlled parameter, such as a subscription level. For example, a medical facility may purchase a subscription to hub connectivity capabilities. Some subscription level(s) may provide the hub access to surgical data gathered from external systems, while others may limit the hub connectivity to internal devices.

The hub connectivity control parameter(s) may include available data bandwidth, power capacity and usage, processor and memory utilization, and/or internal or attached systems.

The hub connectivity control parameter(s) may include an indication from a tiered system. The tiered system may scale the communication between the surgical hub 7006 and the device(s) 7012, the communication between the surgical hub 7006 and external server(s) 7013/7002 and/or the like, based on the available data bandwidth, power capacity and usage, processor and memory utilization, and/or internal or attached systems. The tiered system may determine max communication capabilities the surgical huh may operate under. For example, upon detecting the power capability associated with the operation room, associated with the surgical hub, and/or associated with a medical facility is below a threshold, the tiered system may scale down the surgical hub's connectivity capabilities. For example, upon detecting available data bandwidth is below a threshold, memory utilization is above a certain threshold, power usage is above a certain threshold, and/or other system conditions that may warrant scaling down the surgical hub's connectivity capabilities, the tiered system may limit or disable the communication between the surgical hub and the devices and/or the communication between the surgical hub and external server(s). For example, bi-directional connectivity mode (as shown in FIG. 16B) may be scaled down to flow-through connectivity mode (as shown in FIG. 16A). External communications (as shown in FIG. 16 ) may be disabled. In examples, the tiered system may be a module within the surgical hub or may be a system external to the surgical hub.

At 16110, the surgical hub may communicate with devices in the operating room, servers in the internal and/or external network(s) in accordance with the determined hub connectivity mode.

In an example hub connectivity mode, the surgical hub may receive information from surgical instrument(s) and may send the received information to a remote server (such as a remote processing server and/or a remote database in the cloud).

In an example connectivity mode, the surgical hub may receive information from surgical instrument(s) and may send the received information to a remote server (such as a remote processing server and/or a remote database in the cloud). The surgical hub may receive information from surgical instrument(s), obtain instructional information based on the information received from the surgical instrument(s), and may send the instructional information to one or more surgical instrument(s).

In an example connectivity mode, the surgical hub may receive information from surgical instrument(s) and may send the received information to a remote server (such as a remote processing server and/or a remote database in the cloud). The surgical hub may receive information from surgical instrument(s), obtain instructional information based on the information received from the surgical instrument(s), and may send the instructional information to one or more surgical instrument(s). The surgical hub may record various surgical information and send surgical information to a remote server for archiving and/or analysis. The archived surgical information may be aggregated with information received from other surgical hub(s), and/or surgical information associated with other medical facilities. The aggregated information may be accessed to generate instructional information to one or more surgical instrument(s). In an example, the surgical communication hub may aggregate information, such as information received from smart surgical devices, information associated with multiple surgeries, surgical information and corresponding outcome associated with multiple patients. The aggregated information may be stored in a remote database. In an example, the surgical information may be aggregated at a remote server.

FIGS. 16A-C illustrate example hub connectivity modes that a surgical hub, such as the surgical hub 7006 may operate under. As shown in FIGS. 16A-C, the surgical hub 15504 may communicate with the various devices 15506, remote server(s) in the cloud 15502 and/or devices, servers and databases in external networks 15508 in different connectivity modes.

For example, the surgical hub may determine to operate in a connectivity mode where surgical information may flow through the surgical hub to a remote server. As shown in FIG. 16A, the surgical hub 15504 may serve as a communication portal between the local devices/systems (e.g., surgical instruments and other equipment within the operating room) 15506 and the other connected systems 15502, including systems local or remote to the surgical hub. In this flow-through connectivity mode, the surgical hub 15504 may act as a communication bus between different devices and systems, enabling them to communicate via the surgical hub.

The surgical hub 15504 may receive surgical information data from one or more smart surgical devices 15506 in the operating room, for example, as described herein with reference to FIG. 13 . As shown in FIG. 13 , the surgical hub 9000 may receive surgical data associated with a surgical procedure being performed in the surgical operating room from the modular surgical device(s) 9050.

During a surgical procedure, surgical devices 9050 may track and record surgical data and variables (e.g., surgical parameters). The surgical parameters may include force-to-fire (FTF), force-to-close (FTC), firing progress, tissue gap, power level, impedance, tissue compression stability (creep), and/or the like.

The surgical devices 9050 may include an end effector including a staple cartridge. The captured surgical data may include snapshots taken via an endoscope of the surgical hub during a stapling portion of a surgical procedure. The surgical devices 9050 may include a temperature sensor. The captured surgical data may include least one temperature detected by the temperature sensor during a tissue sealing portion of a surgical procedure.

For example, when operating under flow-through connectivity mode, the surgical hub 15504 may disable interpretation, control or operation on the received information. The surgical hub 15504 may determine whether to disable obtaining instructional information based on the connectivity mode. Based on a determination that the current connectivity mode is a flow-through mode, the surgical hub 15504 may disable obtaining instructional information.

In an example hub connectivity mode, the surgical hub may generate instructional information based on the received surgical data.

FIG. 18A shows an example flow for operating under variable hub communication modes. As shown in FIG. 18A, at 16182, a hub connectivity mode may be determined based on the identified hub connectivity control parameter(s) as described herein. At 16184, whether the hub connectivity mode indicates the hub may provide instructional information to surgical devices may be determined. For example, the surgical hub 5104 as described with respect to FIG. 9 may determine whether to provide instructional information to at least one smart surgical device 5102 based on the hub connectivity mode. On a condition that the hub connectivity mode does not support provisioning instructional information to surgical devices, at 16188, provisioning instructional information to surgical devices may be disabled. On a condition that the hub connectivity mode supports provisioning instructional information to surgical devices, at 16186, the surgical hub may determine to obtain and provide instructional information to surgical devices.

The surgical hub may obtain instructional information to surgical devices based at least in part on surgical data received from one or more surgical devices. For example, based on a determination that the current hub connectivity mode is a bi-directional mode, the surgical hub may receive surgical data from a surgical device and may obtain a response to the surgical device based on the received surgical data. Based on a determination that the current hub connectivity mode is a bi-directional mode, the surgical hub may receive surgical data from a first device and may obtain an indication to a second device based on the surgical data received from the first device. The indication to the second device may include the surgical data received from the first device and/or other information.

FIG. 16B shows an example bi-directional mode. As shown, the surgical hub 15506 may receive surgical data from surgical device(s) 15506, and may send data, such as instructional information to device(s) 15506. The surgical hub 15506 may aggregate the surgical data received from surgical device(s) 15506 prior to sending to the remote server(s) in the cloud 15502.

In an example, the surgical data received by the surgical hub and sent to the remote server(s) may include a property of airborne particles in a fluid within a patient's abdominal cavity, such as a particle type, particle size, particle concentration, particle velocity, and/or particle direction. The instructional information that the surgical hub obtains based on the received surgical data may include, but not limited to, an adjustment to a surgical function, such as proportionately increasing the surgical function based on the property of airborne particles in the fluid, adding a supplemental surgical function to the surgical function, adjusting the power level provided to an energy device, adjusting the speed of a pump in a smoke evacuator, adjusting a flow path through the filtering system of the smoke evacuator, adjusting the operating room vent to increase ventilation therethrough, adjusting a degree of the activation of an actuator, and/or replacing the surgical function with an alternative surgical function.

In examples, the instructional information that the surgical hub obtains based on the received surgical data may include, but not limited to, prioritization information (e.g., display prioritization information), cartridge usage or selection recommendation, a warning message and/or surgical device usage instructions.

For example, when the current hub connectivity mode is a bi-directional mode, the surgical hub may determine to obtain and provide instructional information. An example bi-directional connectivity mode may enable situational awareness and controlling surgical device(s). The surgical hub may infer progression of the surgical procedure from the surgical data and may obtain instructional information based on the inferred progression of the surgical procedure. The surgical hub may assess a surgical activity performed by an end effector of the modular surgical device at the surgical site from the data extracted from the at least one image frame.

For example, as shown in FIG. 9 , the surgical hub 5104 may receive perioperative data from the devices 5102 and other data sources (e.g., databases 5122 and patient monitoring devices 5124) that are communicably coupled to the surgical hub 5706. The surgical hub 5104 may determine whether an event has occurred based on the received data. The event can include, for example, a surgical procedure, a step or portion of a surgical procedure, or downtime between surgical procedures or steps of a surgical procedure. The surgical hub 5104 may track data associated with the particular event, such as the length of time of the event, the surgical instruments and/or other medical products utilized during the course of the event, and the medical personnel associated with the event. The surgical hub 5104 may determine event data via, for example, the situational awareness processes as described herein.

The surgical hub 5104 may provide the devices 5102 with instructional information such as, but not limited to, control adjustment information, prioritization information, warning, display instructions. For example, the surgical hub 5104 may receive surgical data that may include perioperative data detected by one or more smart surgical devices 5102 during a surgical procedure. The surgical hub 5104 may determine contextual information regarding the surgical procedure according to the perioperative data. The surgical hub may obtain control adjustments for one or more surgical devices 5102 based on the contextual information. The perioperative data include one or more parameter associated with the modular device and/or one or more parameter associated with a patient.

The instructional data information be provided to a surgical device with an associated priority. The instructional information that the surgical hub obtains based on the received surgical data may include recommendation to a clinician in an operating room. The recommendation may be provided to a surgical device with a priority that may be determined by the surgical hub. For example, an elevated priority level may be communicated with at least one of marking, emphasizing, highlighting, or flashing. For example, the surgical hub may determine a surgical state based on the received surgical data and may determine the priority level of the recommendation based on the surgical state. For example, the surgical hub 5104 may determine a surgical state based on the received surgical data. The surgical state may include, a step in a surgical procedure, identification of a suite of surgical devices currently in use in a surgical theater, a position of a portion of the surgical device, a position of a jaw of an end effector of the surgical device, a gross usage surgical step, and/or a precision usage surgical step.

The surgical huh may select one or more recommendations from multiple possible recommendations based on the surgical state. The priority level of the recommendation may be adjusted based on an anticipated surgical action. The anticipated surgical action may be determined based on the surgical state, and/or based on a position of a surgical device at a surgical site.

Situational awareness processes are described in greater detail in U.S. patent application Ser. No. 15/940,654, titled SURGICAL HUB SITUATIONAL AWARENESS, filed Mar. 29, 2018; U.S. patent application Ser. No. 16/209,478, titled METHOD FOR SITUATIONAL AWARENESS FOR SURGICAL NETWORK OR SURGICAL NETWORK CONNECTED DEVICE CAPABLE OF ADJUSTING FUNCTION BASED ON A SENSED SITUATION OR USAGE, filed Dec. 4, 2018; and U.S. patent application Ser. No. 16/182,246, titled ADJUSTMENTS BASED ON AIRBORNE PARTICLE PROPERTIES, filed Nov. 6, 2018; the disclosure of each is herein incorporated by reference in its entirety.

Surgical procedures are performed by different surgeons at different locations, some with much less experience than others. For a given surgical procedure, there may be many parameters that can be varied to attempt to realize a desired outcome. For example, for a given surgical procedure which utilizes energy supplied by a generator, the surgeon often relies on experience alone for determining which mode of energy to utilize, which level of output power to utilize, the duration of the application of the energy, etc., in order to attempt to realize the desired outcome. To increase the likelihood of realizing desired outcomes for different surgical procedures, a surgeon may be provided with best practice recommendations, which may be generated based on aggregated surgical data sets associated with multiple surgical procedures performed in multiple locations over time.

As shown in FIG. 18B, at 16202, a hub connectivity mode may be determined based on the identified hub connectivity control parameter(s) as described herein. At 16204, whether the determined hub connectivity mode supports data aggregation with external sources may be determined. For example, the hub may determine whether to send recorded surgical information associated with a procedure to a remote server for archiving and potential aggregation with data associated with external network(s) based on the hub connectivity mode. If the determined hub connectivity mode supports data aggregation with external sources, at 16206, the surgical hub may send recorded surgical information to a remote server. For example, the surgical hub may enable communication to external system(s) when operating in a certain connectivity mode. Based on a determination to send recorded surgical information to a remote server, surgical data may be sent to an external hub associated with an external network (e.g., a different medial facility, a different hospital, or the like).

FIG. 16C shows an example hub connectivity mode that supports data aggregation with external data sets. As shown, the surgical hub 15506 may receive surgical data from surgical device(s) 15506, and may send data, such as instructional information to device(s) 15506. The surgical hub 15504 may facilitate recording and archiving surgical data and may exchange surgical data and/or related analysis with an external network(s) 15508. Data from various hospitals or medical organizations 15508 can be aggregated. Surgical data, outcome, patient information can be compiled to determine instructional information, surgical recommendations, aggregation analysis, and/or the like. As shown, the surgical hub 15504 may retrieve aggregation analysis from remote server(s) or database(s) in the cloud 15502. The aggregation analysis may be used to generate instructional information for sending to surgical devices 15506.

As shown in FIG. 18B, the surgical hub may determine whether to disable communication to an external system based on the hub connectivity mode. If the determined hub connectivity mode does not support data aggregation with external sources, at 16208, sending recorded surgical information for aggregation with external sources may be disabled. For example, the surgical hub may disable communication to external system(s) when operating under certain connectivity modes, such as the flow-through connectivity mode and the bi-directional connectivity mode described herein.

Recording surgical data is described in greater detail in U.S. patent application Ser. No. 16/209,385, titled METHOD OF HUB COMMUNICATION, PROCESSING, STORAGE AND DISPLAY, filed Dec. 4, 2018, the disclosure of which is herein incorporated by reference in its entirety. The recorded surgical data may include surgical event data as described herein. Surgical event data, for example, recorded and/or aggregated surgical event data may be sent to a remote server for aggregation with surgical data from external networks and for further analysis.

Examples of aggregation (e.g., remote aggregation), requests and analysis are described in detail in U.S. patent application Ser. No. 15/940,668 titled AGGREGATION AND REPORTING OF SURGICAL HUB DATA;, filed on Mar. 29, 2018, which is herein incorporated by reference in its entirety.

When operating in a connectivity mode that allows external communication, the surgical hub may request information from a remote server and/or external systems. As shown in FIG. 18C, at 16302, a hub connectivity mode may be determined based on the identified hub connectivity control parameter(s) as described herein. At 16304, the hub may determine whether to retrieve aggregation analysis from the remote server based on the hub connectivity mode. Based on a determination that the current hub connectivity mode supports remote data aggregation and analysis, at 16308, the surgical hub may generate an aggregation analysis request. The request may be generated based on the received surgical data and may be sent to a remote server at 16310. For example, the aggregation analysis request may indicate a request for recommendation on generator data associated with a particular step in a surgical procedure. In response, the surgical hub may receive an aggregation analysis response from the remote server at 16312.

For example, the aggregation analysis response may include a recommendation and/or a report. The aggregation analysis response may include one or more of: an energy mode of the generator for a particular surgical procedure, a power output of the generator for a particular surgical procedure, and/or a duration of the power output of the generator for a particular surgical procedure. The aggregation analysis response may include instructional information as described herein. At 16314, the surgical hub may generate and send instructional information to one or more surgical device(s) based on the received aggregation analysis response. As shown in FIG. 18C, based on a determination that the current hub connectivity mode supports remote data aggregation analysis, the surgical hub may disable data aggregation analysis requests at 16306.

FIG. 19 is a block diagram of a computer-implemented interactive surgical system 5700, in accordance with at least one aspect of the present disclosure. The system 5700 may include a number of surgical hubs 5706 that, as described herein, may detect and track data related to surgical procedures that the surgical hubs 5706 (and the modular devices paired to the surgical hubs 5706) are utilized in connection with. The surgical hubs 5706 may be connected to form local networks such that the data being tracked by the surgical hubs 5706 is aggregated together across the network. The networks of surgical hubs 5706 can be associated with a medical facility, for example. The data aggregated from the network of surgical hubs 5706 can be analyzed to provide reports on data trends or recommendations. For example, the surgical hubs 5706 of a first medical facility 5704 a may be communicably connected to a first local database 5708 a and the surgical hubs 5706 of a second medical facility 5704 b are communicably connected to a second local database 5708 b. The network of surgical hubs 5706 associated with the first medical facility 5704 a can be distinct from the network of surgical hubs 5706 associated with the second medical facility 5704 b, such that the aggregated data from networks of surgical hubs 5706 may correspond to medical facility 5704 a, 5704 b individually. A surgical hub 5706 or another computer terminal communicably connected to the database 5708 a, 5708 b can be configured to provide reports or recommendations based on the aggregated data associated with the respective medical facility 5704 a, 5704 b. The data tracked by the surgical hubs 5706 can be utilized to, for example, report whether a particular incidence of a surgical procedure deviated from the average in-network time to complete the particular procedure type.

The surgical hub 5706 may upload the tracked data to the cloud 5702, for processing and aggregating the tracked data across multiple surgical hubs 5706, networks of surgical hubs 5706, and/or medical facilities 5704 a, 5704 b that are connected to the cloud 5702. The surgical hub 5706 may provide reports or recommendations based on the aggregated data. The data tracked by the surgical hubs 5706 can be utilized to, for example, report whether a particular incidence of a surgical procedure deviated from the average global time to complete the particular procedure type.

The surgical hub 5706 can be configured to access the cloud 5702 to compare locally tracked data to global data aggregated from the surgical hubs 5706 that are communicably connected to the cloud 5702. The surgical hub 5706 may provide reports or recommendations based on the comparison between the tracked local data relative to local (e.g., in-network) or global norms. The data tracked by the surgical hubs 5706 can be utilized to, for example, report whether a particular incidence of a surgical procedure deviated from either the average in-network time or the average global time to complete the particular procedure type.

The surgical huh 5706 or a computer system local to the surgical hub 5706 may locally aggregate the data tracked by the surgical hubs 5706, store the tracked data, and generate reports and/or recommendations according to the tracked data in response to queries. In cases where the surgical hub 5706 is connected to a medical facility network (which may include additional surgical hubs 5706), the surgical hub 5706 may compare the tracked data with the bulk medical facility data. The bulk medical facility data can include EMR data and aggregated data from the local network of surgical hubs 5706. The cloud 5702 (e.g., a remote server in the cloud) may aggregate the data tracked by the surgical hubs 5706, store the tracked data, and generate reports and/or recommendations according to the tracked data in response to queries.

The surgical hub 5706 can provide reports regarding trends in the data and/or provide recommendations on improving the efficiency or effectiveness of the surgical procedures being performed. The data trends and recommendations can be based on data tracked by the surgical hub 5706 itself, data tracked across a local medical facility network containing multiple surgical hubs 5706, and/or data tracked across a number of surgical hubs 5706 communicably connected to a cloud 5702. The recommendations provided by the surgical hub 5706 can describe, for example, particular surgical instruments or product mixes to utilize for particular surgical procedures based on correlations between the surgical instruments/product mixes and patient outcomes and procedural efficiency. The reports provided by the surgical hub 5706 can describe, for example, whether a particular surgical procedure was performed efficiently relative to local or global norms, whether a particular type of surgical procedure being performed at the medical facility is being performed efficiently relative to global norms, and the average time taken to complete a particular surgical procedure or step of a surgical procedure for a particular surgical team.

The surgical hub 5706 may determine when operating theater events occur (e.g., via a situational awareness module/system) and track the length of time spent on each event. An operating theater event is an event that a surgical hub 5706 can detect or infer the occurrence of. An operating theater event can include, for example, a particular surgical procedure, a step or portion of a surgical procedure, or downtime between surgical procedures. The operating theater events can be categorized according to an event type, such as a type of surgical procedure being performed, so that the data from individual procedures can be aggregated together to form searchable data sets. The data tracked by the surgical hubs 5706 being parsed to provide metrics related to surgical procedures or the use of the surgical hub 5706.

The surgical hub 5706 may determine whether a surgical procedure is being performed and then track both the length of time spent between procedures (e.g., downtime) and the time spent on the procedures themselves. The surgical hub 5706 can determine and track the time spent on the individual steps taken by the medical personnel (e.g., surgeons, nurses, orderlies) either between or during the surgical procedures. The surgical hub may determine when surgical procedures or different steps of surgical procedures are being performed via a situational awareness module/system as described in herein.

Near-field-communication (NFC) cards may be used to automate supply chain. FIG. 20 illustrates an example surgical supply packaged with a radio frequency identification (RFID) NFC chip. As shown, surgical supply 16510 may include surgical devices, sutures, biosurgery supply, and/or the like. Medical personnel may scan the chip 16530 on the surgical supply 16510 prior to introduction to surgery. This may allow the supply chain for customers ordering and may provide full case device profiles, for example, via cloud 16520. NFC reader may be used to track inventory, for example, by adding an RFID card 16530 into devices or packaging.

Examples of the Disclosure

The following is a non-exhaustive list of embodiments described hereinabove and/or shown in the drawings, and which may be claimed hereinafter:

Example 1. A surgical hub comprising:

-   -   a receiver configured to: receive surgical information from at         least one smart surgical device;     -   a transmitter configured to: send surgical data to a remote         server;     -   a processor configured to:         -   determine a huh connectivity mode based on a huh             connectivity control parameter;         -   determine whether to send instructional information to the             at least one smart device based on the hub connectivity             mode; and         -   communicate with the at least one smart device based on the             determination.

The surgical hub according to Example 1 may be the surgical hub as shown in FIG. 11 and described above, though alternative configurations of surgical hub are also envisaged. The surgical hub may operate under one of the exemplary hub connectivity modes shown in FIGS. 16A-C, though alternative connectivity modes are also envisaged. Irrespective of the particular configuration, a surgical hub may communicate with various devices, remote server(s), and/or devices, servers and databases in external networks in different connectivity modes. The communication allows the hub to communicate to and from these devices, aggregate information and then communicate to and from a remote processing server or database.

The hub connectivity mode recited in Example 1 may be selected from multiple connectivity modes that may be preconfigured, dynamically updated, semi-dynamically updated, periodically updated, or preset. The hub connectivity modes may control inter-device connectivity within a network associated with a medical facility, e.g. a hospital, and/or communication with an external network associated with a different hospital. In Example 1, the hub connectivity mode is determined based on a hub connectivity control parameter.

The hub connectivity control parameter(s) may include, but not limited to, systems capabilities such as hardware capability, firmware capability and/or software capability. The hub connectivity control parameter(s) may include a consumer-controlled parameter, such as a subscription level. The hub connectivity control parameter(s) may include an indication from a tiered system. The tiered system may scale the communication between the surgical hub and the device(s), the communication between the surgical hub and external server(s) and/or the like, based on the available data bandwidth, power capacity and usage, processor and memory utilization, and/or internal or attached systems. The tiered system may determine max communication capabilities the surgical hub may operate under.

The surgical hub of Example 1 may operate in a flow-through connectivity mode, where surgical information may flow through the surgical hub to a remote server. The surgical hub may operate in a bi-directional mode in which the surgical hub may receive surgical data from surgical device(s) and send data, such as instructional information to device(s). In an exemplary bi-directional mode, the surgical hub may aggregate the surgical data received from surgical device(s) prior to sending to the remote server(s) in the cloud. The bi-directional mode may enable situational awareness and controlling surgical device(s). The surgical hub may operate in a hub connectivity mode that supports data aggregation with external data sets. In this hub connectivity mode, the surgical hub may facilitate recording and archiving surgical data and may exchange surgical data and/or related analysis with an external network(s). Data from various hospitals or medical organizations can be aggregated. Surgical data, outcome, patient information can be compiled to determine instructional information, surgical recommendations, aggregation analysis, and/or the like. The surgical hub may retrieve aggregation analysis from remote server(s) or database(s) in a cloud. Aggregation analysis increases the likelihood of realizing desired outcomes for different surgical procedures. A surgeon may be provided with best practice recommendations, which may be generated based on aggregated surgical data sets associated with multiple surgical procedures performed in multiple locations over time. 

The invention claimed is:
 1. A surgical hub comprising: a receiver configured to: receive surgical information from at least one smart surgical device; a transmitter configured to: send surgical data to a remote server; a processor configured to: determine a hub connectivity control parameter, the hub connectivity control parameter indicating at least one of hardware capability, firmware capability, or software capability; select, based on the hub connectivity control parameter, a hub connectivity mode from a plurality of hub connectivity modes, the hub connectivity mode for controlling connectivity with the surgical hub; determine whether to send instructional information to the at least one smart surgical device based on the hub connectivity mode; and communicate with the at least one smart surgical device based on a determination to send instructional information.
 2. The surgical hub of claim 1, wherein the transmitter is further configured to send data to the at least one smart surgical device, and wherein the processor is further configured to: based on a determination to send instructional information, obtain instructional information based on the received surgical information; and send the obtained instructional information to the at least one smart surgical device via the transmitter.
 3. The surgical hub of claim 1, wherein the instructional information comprises at least one of: an adjustment to a surgical function; prioritization information; a cartridge usage recommendation; a warning message; surgical device usage recommendations; or surgical device usage instructions.
 4. The surgical hub of claim 1, wherein the processor is further configured to: determine whether to disable communication with an external system based on the hub connectivity mode.
 5. The surgical hub of claim 1, wherein the processor is further configured to: determine whether to send recorded surgical information associated with a procedure to the remote server based on the hub connectivity mode.
 6. The surgical hub of claim 1, wherein the receiver is further configured to receive data from the remote server, and wherein the processor is further configured to: determine whether to retrieve aggregation analysis from the remote server based on the hub connectivity mode; based on a determination to retrieve the aggregation analysis, generate an aggregation analysis request based on the received surgical information; send the aggregation analysis request to the remote server via the transmitter; receive an aggregation analysis response from the remote server via the receiver; generate the instructional information based on the aggregation analysis response; and send the instructional information to the at least one smart surgical device via the transmitter.
 7. The surgical hub of claim 1, wherein the hub connectivity control parameter comprises at least one of: available data bandwidth, power capacity associated with the surgical hub, power capacity associated with an operating room, power capacity associated with a medical facility, a power usage, processor utilization information, or memory utilization information.
 8. The surgical hub of claim 1, wherein the hub connectivity control parameter comprises at least one of: a subscription level associated with hub connectivity, a user preference associated with hub connectivity, or an indication from a tiered software control system.
 9. A method for controlling hub connectivity, the method comprising: receiving, by a surgical hub, surgical information from at least one smart surgical device; determining, by the surgical hub, a hub connectivity control parameter, the hub connectivity control parameter indicating at least one of hardware capability, firmware capability, or software capability; selecting, by the surgical hub, based on the hub connectivity control parameter, a hub connectivity mode from a plurality of preconfigured hub connectivity modes, the hub connectivity mode for controlling connectivity with the surgical hub; determining, by the surgical hub, whether to send instructional information to the at least one smart surgical device based on the hub connectivity mode; and communicating, by the surgical hub, with the at least one smart surgical device based on a determination to send instructional information.
 10. The method of claim 9, further comprising: based on a determination to send instructional information, obtaining instructional information based on the received surgical information; and sending the obtained instructional information to the at least one smart surgical device.
 11. The method of claim 9, wherein the instructional information comprises at least one of: an adjustment to a surgical function; prioritization information; a cartridge usage recommendation; a warning message; surgical device usage recommendations; or surgical device usage instructions.
 12. The method of claim 9, further comprising: determining whether to disable communication with an external system based on the hub connectivity mode.
 13. The method of claim 9, further comprising: determining whether to send recorded surgical information associated with a procedure to a remote server based on the hub connectivity mode.
 14. The method of claim 9, further comprising: receiving data from the remote server; determining whether to retrieve aggregation analysis from the remote server based on the hub connectivity mode; based on a determination to retrieve the aggregation analysis, generating an aggregation analysis request based on the received surgical information; sending the aggregation analysis request to the remote server via a transmitter; receiving an aggregation analysis response from the remote server via the receiver; generating the instructional information based on the aggregation analysis response; and sending the instructional information to the at least one smart surgical device via the transmitter.
 15. The method of claim 9, wherein the hub connectivity control parameter comprises at least one of: available data bandwidth, power capacity associated with the surgical hub, power capacity associated with an operating room, power capacity associated with a medical facility, a power usage, processor utilization information, or memory utilization information.
 16. The method of claim 9, wherein the hub connectivity control parameter comprises at least one of: a subscription level associated with hub connectivity, a user preference associated with hub connectivity, or an indication from a tiered software control system.
 17. A non-transitory computer readable medium comprising instructions stored therein, when executed performing: receiving, by a surgical hub, surgical information from at least one smart surgical device; determining, by the surgical hub, a hub connectivity control parameter, the hub connectivity control parameter indicating at least one of hardware capability, firmware capability, or software capability; selecting, by the surgical hub, based on the hub connectivity control parameter, a hub connectivity mode from a plurality of preconfigured hub connectivity modes, the hub connectivity mode for controlling connectivity with the surgical hub; determining, by the surgical hub, whether to send instructional information to the at least one smart surgical device based on the hub connectivity mode; and communicating, by the surgical hub, with the at least one smart surgical device based on a determination.
 18. The non-transitory computer readable medium of claim 17, further comprising instructions when executed performing: based on a determination to send instructional information, obtaining instructional information based on the received surgical information; and sending the obtained instructional information to the at least one smart surgical device.
 19. The non-transitory computer readable medium of claim 17, further comprising instructions when executed performing: determining whether to disable communication with an external system based on the hub connectivity mode.
 20. The non-transitory computer readable medium of claim 17, further comprising instructions when executed performing: determining whether to send recorded surgical information associated with a procedure to a remote server based on the hub connectivity mode. 